| ||Format||Pages||Price|| |
|15||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||15||$52.00||  ADD TO CART|
|Standard + Redline PDF Bundle||30||$62.40||  ADD TO CART|
Significance and Use
5.1 Pose measurement systems are used in a wide range of fields including manufacturing, material handling, construction, medicine, and aerospace. The use of pose measurement systems could, for example, replace the need to fix the poses of objects of interest by mechanical means.
5.2 Potential users have difficulty comparing pose measurement systems because of the lack of standard performance specifications and test methods, and must rely on the specifications of a vendor regarding the system’s performance, capabilities, and suitability for a particular application. This standard makes it possible for a user to assess and compare the performance of candidate pose measurement systems, and allows the user to determine if the measured performance results are within the vendor’s claimed specifications with regard to the user’s application. This standard also facilitates the improvement of pose measurement systems by providing a common set of metrics to evaluate system performance.
5.3 The intent of this test method is to allow a user to determine the performance of a vendor’s system under conditions specific to the user’s application, and to determine whether the system still performs in accordance with the vendor’s specifications under those conditions. The intention of this test method is not to validate a vendor’s claims; although, under specific situations, this test method may be adapted for this purpose.
1.1 Purpose—In this test method, metrics and procedures for collecting and analyzing data to determine the performance of a pose measurement system in computing the pose (position and orientation) of a rigid object are provided.
1.2 This test method applies to the situation in which both the object and the pose measurement system are static with respect to each other when measurements are performed. Vendors may use this test method to establish the performance limits for their six degrees of freedom (6DOF) pose measurement systems. The vendor may use the procedures described in to generate the test statistics, then apply an appropriate margin or scaling factor as desired to generate the performance specifications. This test method also provides a uniform way to report the relative or absolute pose measurement capability of the system, or both, making it possible to compare the performance of different systems.
1.3 Test Location—The methodology defined in this test method shall be performed in a facility in which the environmental conditions are within the pose measurement system’s rated conditions and meet the user’s requirements.
1.4 Units—The values stated in SI units are to be regarded as the standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E456 Terminology Relating to Quality and Statistics
E2544 Terminology for Three-Dimensional (3D) Imaging Systems
ASME StandardASME B89.4.19 Performance Evaluation of Laser-Based Spherical Coordinate Measurement Systems
ISO/IEC StandardsIEC 60050-300:2001 International Electrotechnical Vocabulary--Electrical and Electronic Measurements and Measuring Instruments JCGM 100:2008 Evaluation of Measurement Data--Guide to the Expression of Uncertainty in Measurement (GUM) JCGM 200:2012 International Vocabulary of Metrology--Basic and General Concepts and Associated Terms (VIM), 3rd edition
ICS Number Code 17.040.99 (Other standards relating to linear and angular measurements)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2919-14, Standard Test Method for Evaluating the Performance of Systems that Measure Static, Six Degrees of Freedom (6DOF), Pose, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top