| ||Format||Pages||Price|| |
|13||$58.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$58.00||  ADD TO CART|
|Standard + Redline PDF Bundle||26||$69.00||  ADD TO CART|
Significance and Use
4.1 The major factors affecting the quality of a CT image are total image unsharpness (UTimage), contrast (Δµ), and random noise (σ). Geometrical and detector unsharpness limit the spatial resolution of a CT system, that is, its ability to image fine structural detail in an object. Random noise and contrast response limit the contrast sensitivity of a CT system, that is, its ability to detect the presence or absence of features in an object. Spatial resolution and contrast sensitivity may be measured in various ways. In this test method, spatial resolution is quantified in terms of the modulation transfer function (MTF), and contrast sensitivity is quantified in terms of the contrast discrimination function (CDF). The relationship between contrast sensitivity and spatial resolution describing the resolving and detecting capabilities is given by the contrast-detail-diagram (CDD metric, see also Guide and Practice ). This test method allows the purchaser or the provider of CT systems or services, or both, to measure and specify spatial resolution and contrast sensitivity and is a measure for system stability over time and performance acceptability.
1.1 This test method provides instruction for determining the spatial resolution and contrast sensitivity in X-ray and γ-ray computed tomography (CT) volumes. The determination is based on examination of the CT volume of a uniform cylinder of material. The spatial resolution measurement (Modulation Transfer Function) is derived from an image analysis of the sharpness at the edges of the reconstructed cylinder slices. The contrast sensitivity measurement (Contrast Discrimination Function) is derived from an image analysis of the contrast and the statistical noise at the center of the cylinder slices.
1.2 This test method is more quantitative and less susceptible to interpretation than alternative approaches because the required cylinder is easy to fabricate and the analysis easy to perform.
1.3 This test method is not to predict the detectability of specific object features or flaws in a specific application. This is subject of IQI and RQI standards and standard practices.
1.4 This method tests and describes overall CT system performance. Performance tests of systems components such as X-ray tubes, gamma sources, and detectors are covered by separate documents, namely Guide , Practice , and Practice ; c.f. , which should be consulted for further system analysis.
1.5 Units—The values stated in SI units are to be regarded as standard. The values given in parentheses after SI units are provided for information only and are not considered standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E177 Practice for Use of the Terms Precision and Bias in ASTM Test Methods
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E1000 Guide for Radioscopy
E1316 Terminology for Nondestructive Examinations
E1441 Guide for Computed Tomography (CT)
E1570 Practice for Fan Beam Computed Tomographic (CT) Examination
E2002 Practice for Determining Total Image Unsharpness and Basic Spatial Resolution in Radiography and Radioscopy
E2737 Practice for Digital Detector Array Performance Evaluation and Long-Term Stability
ISO Standard15708 NDT Radiation Methods Computed Tomography Part 1: Terminology, Part 2: Principles, Equipment and Samples, Part 3: Operation and Interpretation, Part 4: Qualification
ICS Number Code 11.040.50 (Radiographic equipment)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E1695-20, Standard Test Method for Measurement of Computed Tomography (CT) System Performance, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top