| ||Format||Pages||Price|| |
|7||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||7||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||14||$54.00||  ADD TO CART|
Significance and Use
5.1 The pour point of a petroleum specimen is an index of the lowest temperature of its utility for certain applications.
1.1 This test method covers and is intended for use on any petroleum product. A procedure suitable for black specimens, cylinder stock, and nondistillate fuel oil is described in . The cloud point procedure formerly part of this test method now appears as Test Method .
1.2 Currently there is no ASTM test method for automated Test Method pour point measurements.
1.3 Several ASTM test methods offering alternative procedures for determining pour points using automatic apparatus are available. None of them share the same designation number as Test Method . When an automatic instrument is used, the ASTM test method designation number specific to the technique shall be reported with the results. A procedure for testing the pour point of crude oils is described in Test Method .
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 WARNING—Mercury has been designated by many regulatory agencies as a hazardous material that can cause central nervous system, kidney and liver damage. Mercury, or its vapor, may be hazardous to health and corrosive to materials. Caution should be taken when handling mercury and mercury containing products. See the applicable product Material Safety Data Sheet (MSDS) for details and EPA’s website—http://www.epa.gov/mercury/faq.htm—for additional information. Users should be aware that selling mercury and/or mercury containing products into your state or country may be prohibited by law.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D117 Guide for Sampling, Test Methods, and Specifications for Electrical Insulating Oils of Petroleum Origin
D396 Specification for Fuel Oils
D2500 Test Method for Cloud Point of Petroleum Products
D5853 Test Method for Pour Point of Crude Oils
D6300 Practice for Determination of Precision and Bias Data for Use in Test Methods for Petroleum Products and Lubricants
D7962 Practice for Determination of Minimum Immersion Depth and Assessment of Temperature Sensor Measurement Drift
E1 Specification for ASTM Liquid-in-Glass Thermometers
E644 Test Methods for Testing Industrial Resistance Thermometers
E1137 Specification for Industrial Platinum Resistance Thermometers
E2877 Guide for Digital Contact Thermometers
ICS Number Code 75.080 (Petroleum products in general)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D97-17a, Standard Test Method for Pour Point of Petroleum Products, ASTM International, West Conshohocken, PA, 2017, www.astm.orgBack to Top