| ||Format||Pages||Price|| |
|11||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||11||$54.00||  ADD TO CART|
|Standard + Redline PDF Bundle||22||$65.00||  ADD TO CART|
Significance and Use
5.1 The test method is useful for the determination of element concentrations in many natural waters, metallurgical process cyanide solutions and wastewaters. It has the capability for the determination of up to 21 elements. High analysis sensitivity can be achieved for some elements that are difficult to determine by other techniques.
1.1 This test method covers the determination of dissolved elements in ground water, surface water, and drinking water. It may also be used for the determination of total-recoverable elements in these waters as well as wastewater.
1.2 This test method should be used by analysts experienced in the use of inductively coupled plasma—mass spectrometry (ICP-MS), the interpretation of spectral and matrix interferences and procedures for their correction.
1.3 It is the user's responsibility to ensure the validity of the test method for waters of untested matrices.
1.4 lists elements for which the test method applies, with recommended masses and typical estimated instrumental detection limits using conventional pneumatic nebulization. Actual working detection limits are sample dependent and, as the sample matrix varies, these detection limits may also vary. In time, other elements may be added as more information becomes available and as required.
1.4.1 This method covers the analysis of mine dewatering groundwater and wastewater effluent in the range of 2–120 μg/L dissolved antimony and 3–200 μg/L dissolved arsenic.
1.4.2 This method covers the analysis of metallurgical processing cyanide solutions in the range of 1–500 µg/L dissolved gold.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1066 Practice for Sampling Steam
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D3370 Practices for Sampling Water from Closed Conduits
D5810 Guide for Spiking into Aqueous Samples
D5847 Practice for Writing Quality Control Specifications for Standard Test Methods for Water Analysis
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E1601 Practice for Conducting an Interlaboratory Study to Evaluate the Performance of an Analytical Method
E1763 Guide for Interpretation and Use of Results from Interlaboratory Testing of Chemical Analysis Methods
ICS Number Code 13.060.50 (Examination of water for chemical substances)
UNSPSC Code 41115404(Mass spectrometers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D5673-16, Standard Test Method for Elements in Water by Inductively Coupled Plasma—Mass Spectrometry, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top