Active Standard ASTM E2911 | Developed by Subcommittee: E13.08
Book of Standards Volume: 03.06
Significance and Use
4.1 Generally, Raman spectra measured using grating-based dispersive or Fourier transform Raman spectrometers have not been corrected for the instrumental response (spectral responsivity of the detection system). Raman spectra obtained with different instruments may show significant variations in the measured relative peak intensities of a sample compound. This is mainly as a result of differences in their wavelength-dependent optical transmission and detector efficiencies. These variations can be particularly large when widely different laser excitation wavelengths are used, but can occur when the same laser excitation is used and spectra of the same compound are compared between instruments. This is illustrated in Fig. 1, which shows the uncorrected luminescence spectrum of SRM 2241, acquired upon four different commercially available Raman spectrometers operating with 785 nm laser excitation. Instrumental response variations can also occur on the same instrument after a component change or service work has been performed. Each spectrometer, due to its unique combination of filters, grating, collection optics and detector response, has a very unique spectral response. The spectrometer dependent spectral response will of course also affect the shape of Raman spectra acquired upon these systems. The shape of this response is not to be construed as either “good or bad” but is the result of design considerations by the spectrometer manufacturer. For instance, as shown in Fig. 1, spectral coverage can vary considerably between spectrometer systems. This is typically a deliberate tradeoff in spectrometer design, where spectral coverage is sacrificed for enhanced spectral resolution.
1.1 This guide is designed to enable the user to correct a Raman spectrometer for its relative spectral-intensity response function using NIST Standard Reference Materials2 in the 224X series (currently SRMs 2241, 2242, 2243, 2244, 2245, 2246), or a calibrated irradiance source. This relative intensity correction procedure will enable the intercomparison of Raman spectra acquired from differing instruments, excitation wavelengths, and laboratories.
ICS Number Code 71.040.50 (Physicochemical methods of analysis)