| ||Format||Pages||Price|| |
|6||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$44.00||  ADD TO CART|
Significance and Use
5.1 Oxidative induction time is a relative measure of the degree of oxidative stability of the material evaluated at the isothermal temperature of the test. The presence, quantity or effectiveness of antioxidants may be determined by this method. The OIT values thus obtained may be compared from one hydrocarbon to another or to a reference material to obtain relative oxidative stability information.
5.2 Typical uses include the oxidative stability of edible oils and fats (oxidative rancidity), lubricants, greases, and polyolefins.
1.1 These test methods describe the determination of the oxidative properties of hydrocarbons by differential scanning calorimetry or pressure differential scanning calorimetry and is applicable to hydrocarbons that oxidize exothermically in their analyzed form.
1.2 Test Method A—A differential scanning calorimeter (DSC) is used at ambient pressure, for example, about 100 kPa of oxygen.
1.3 Test Method B—A pressure DSC (PDSC) is used at high pressure, for example, 3.5 MPa (500 psig) oxygen.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 These test methods are related to ISO 11357–6 but is different in technical content. These test methods are related to CEC L-85–T but includes additional experimental conditions.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Specific precautionary statements are given in and .
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D3350 Specification for Polyethylene Plastics Pipe and Fittings Materials
D3895 Test Method for Oxidative-Induction Time of Polyolefins by Differential Scanning Calorimetry
D4565 Test Methods for Physical and Environmental Performance Properties of Insulations and Jackets for Telecommunications Wire and Cable
D5482 Test Method for Vapor Pressure of Petroleum Products (Mini Method--Atmospheric)
ASTM E1858-08(2015)e1, Standard Test Methods for Determining Oxidation Induction Time of Hydrocarbons by Differential Scanning Calorimetry, ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top