Significance and Use
5.1 This test method is intended for the determination of chromium, bromine, cadmium, mercury, and lead, in homogeneous polymeric materials. The test method may be used to ascertain the conformance of the product under test to manufacturing specifications. Typical time for a measurement is 5 to 10 min per specimen, depending on the specimen matrix and the capabilities of the EDXRF spectrometer.
1.1 This test method describes an energy dispersive X-ray fluorescence (EDXRF) spectrometric procedure for identification and quantification of chromium, bromine, cadmium, mercury, and lead in polymeric materials.
1.2 This test method is not applicable to determine total concentrations of polybrominated biphenyls (PBB), polybrominated diphenyl ethers (PBDE) or hexavalent chromium. This test method cannot be used to determine the valence states of atoms or ions.
1.3 This test method is applicable for a range from 20 mg/kg to approximately 1 wt % for chromium, bromine, cadmium, mercury, and lead in polymeric materials.
1.4 This test method is applicable for homogeneous polymeric material.
1.5 The values stated in SI units are to be regarded as the standard. Values given in parentheses are for information only.
1.6 This test method is not applicable to quantitative determinations for specimens with one or more surface coatings present on the analyzed surface; however, qualitative information may be obtained. In addition, specimens less than infinitely thick for the measured X rays, must not be coated on the reverse side or mounted on a substrate.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.