Significance and Use
5.1 Turbidity is undesirable in drinking water, plant effluent waters, water for food and beverage processing, and for a large number of other water-dependent manufacturing processes. Removal of suspended matter is accomplished by coagulation, settling, and filtration. Measurement of turbidity provides a rapid means of process control to determine when, how, and to what extent the water must be treated to meet specifications.
5.2 This test method is suitable for the on-line monitoring of turbidity such as that found in drinking water, process water, and high purity industrial waters.
5.3 The instrumentation used must allow for the continuous on-line monitoring of a sample stream.
5.4 When reporting the measured result, appropriate units should also be reported. The units are reflective of the technology used to generate the result, and if necessary, provide more adequate comparison to historical data sets.
5.4.1 Table 1 describes technologies and reporting results. Those technologies listed are appropriate for the range of measurement prescribed in this test method are mentioned, though others may come available.
5.4.2 For a specific design that falls outside of the reporting ranges in Table 1, the turbidity should be reported in turbidity units (TU) with a subscripted wavelength value to characterize the light source that was used.
1.1 This test method is applicable to the on-line measurement of turbidity under 5 nephelometric turbidity units (NTU) in water.
1.2 It is the user's responsibility to ensure the validity of this test method for waters of untested matrices.
1.3 In this test method calibration standards are defined in NTU values, but other assigned turbidity units are assumed to be equivalent.
1.4 This test method assigns traceable reporting units to the type of respective technology that was used to perform the measurement. Units are numerically equivalent with respect to the calibration standard. For example, a 1 NTU formazin standard is also equal to a 1 FNU (formazin nephelometric units) standard, a 1 FNRU (formazin nephelometric ratio units) standard, and so forth.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.