| ||Format||Pages||Price|| |
|8||$50.00||  ADD TO CART|
|Hardcopy (shipping and handling)||8||$50.00||  ADD TO CART|
Significance and Use
5.1 Facet Prosthesis Components—The facet replacement may comprise a variety of shapes and configurations. Its forms may include, but are not limited to, ball and socket articulating joints, joints having a free-floating or semi-constrained third body, metallic load-bearing surfaces, and spring and dampening mechanisms. Additionally, it may have a unilateral or bilateral design.
5.2 These test methods are designed to quantify the static and dynamic characteristics of different designs of FPs. The tests are conducted in vitro in order to allow for analysis of individual devices and comparison of the mechanical performance of multiple designs.
5.3 The loads applied to the FP may differ from the complex loading seen in vivo, and therefore, the results from these tests may not directly predict in vivo performance. The results, however, can be used to compare mechanical performance in different devices.
5.4 Fatigue testing in a simulated body fluid or saline may cause fretting, corrosion, or lubricate the interconnections and thereby affect the relative performance of tested devices. This test should be conducted in a 0.9 % saline environmental bath at 37°C at a maximum rate of 10 Hz for metallic devices and 2 Hz for non-metallic devices. Other test environments such as a simulated body fluid, a saline drip or mist, distilled water, other type of lubrication or dry could also be used with adequate justification. Likewise, alternative test frequencies may be used with adequate justification to ensure that they do not impact the device performance.
5.5 It is well known that the failure of materials is dependent upon stress, test frequency, surface treatments, and environmental factors. Therefore, when determining the effect of changing these parameters (for example, frequency, material, or environment), care should be taken to allow for appropriate interpretation of the results. In particular, it may be necessary to assess the influence of test frequency on device fracture while holding the test environment, implant materials and processing, and implant geometry constant.
1.1 This practice provides guidance for the static and dynamic testing of Lumbar Total Facet Prostheses (FPs). These implants are intended to allow motion and lend support to one or more functional spinal unit(s) through replacement of the natural facets.
1.2 These test methods are intended to provide a basis for the mechanical comparison among past, present, and future non-biologic FPs. These test methods allow comparison of devices with different methods of application to the lumbar spine. These test methods are intended to enable the user to mechanically compare devices and do not purport to provide performance standards for them.
1.3 These test methods describe static and dynamic tests by specifying load types and specific methods of applying these loads.
1.4 These test methods do not purport to address all clinically relevant failure modes for FPs, some of which will be device-specific. For example, these test methods do not address implant wear resistance under expected in vivo loads and motions. In addition, the biologic response to wear debris is not addressed in these test methods.
1.5 Requirements are established for measuring displacements and evaluating the stiffness of an FP.
1.6 Some devices may not be testable in all test configurations.
1.7 The values stated in SI units are to be regarded as the standard with the exception of angular measurements, which may be reported in terms of either degrees or radians.
1.8 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.9 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D638 Test Method for Tensile Properties of Plastics
E4 Practices for Force Verification of Testing Machines
E6 Terminology Relating to Methods of Mechanical Testing
E468 Practice for Presentation of Constant Amplitude Fatigue Test Results for Metallic Materials
E739 Practice for Statistical Analysis of Linear or Linearized Stress-Life (S-N) and Strain-Life (-N) Fatigue Data
F1582 Terminology Relating to Spinal Implants
ICS Number Code 11.040.40 (Implants for surgery, prothetics and orthotics)
UNSPSC Code 42321600(Spinal implants)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F2790-10(2019)e1, Standard Practice for Static and Dynamic Characterization of Motion Preserving Lumbar Total Facet Prostheses, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top