| ||Format||Pages||Price|| |
|3||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$44.00||  ADD TO CART|
|Standard + Redline PDF Bundle||6||$52.00||  ADD TO CART|
Significance and Use
5.1 The compatibility of packaging materials with a medical device is a requirement of many regulatory bodies. Since most medical devices are used or implanted in, around, or on the human body, the benefits of these devices must outweigh the risks. Therefore, the packaging materials that come in contact with the medical device must also be evaluated and determined to be safe for use with the human body in that they have no negative impact on the physical, chemical, or biological properties of the device. This evaluation may include both a study of relevant experience with, and actual testing of, packaging materials. Such an evaluation may result in the conclusion that no testing is needed if the material has a demonstrable history of safe use in the specific role that is the same as that of the package under design.
5.2 The medical device manufacturer determines the need for appropriate testing, with consideration of the device/package interactions, if any. When screening information is needed regarding the biocompatibility of the packaging, cytotoxicity testing from the supplier is typically performed.
1.1 This guide provides information to determine the appropriate testing for biocompatibility of medical device packaging materials that have the potential to contact the patient directly or indirectly.
1.2 This guide does not apply to secondary or tertiary packaging materials.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
F17 Terminology Relating to Primary Barrier Packaging
ICS Number Code 11.120.99 (Other standards related to pharmaceutics); 55.020 (Packaging and distibution of goods in general)
UNSPSC Code 24120000(Packaging materials)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F2475-20, Standard Guide for Biocompatibility Evaluation of Medical Device Packaging Materials, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top