| ||Format||Pages||Price|| |
|3||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$44.00||  ADD TO CART|
This specification covers polyoxymethylene (acetal) resin for medical applications. This specification provides requirements and associated test methods for a form of this thermoplastic which is intended for use in manufacturing medical devices, instrumentation or components thereof. Physical and mechanical properties of medical grade polyoxymethylene such as specific gravity, water absorption, equilibrium, tensile yield strength, tensile elongation break, tensile modulus, tensile impact strength, compressive strength, and sheer strength shall be determined. Biocompatibility of acetal resins and implant devices made using these materials shall also be determined.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers polyoxymethylene (acetal) resin for medical applications. This specification provides requirements and associated test methods for a form of this thermoplastic which is intended for use in manufacturing medical devices, instrumentation or components thereof.
1.2 As with any material, some characteristics may be altered by the processing techniques (such as molding, extrusion, machining, sterilization, and so forth) required for a specific application. Therefore properties of fabricated forms of this resin should be evaluated using appropriate test methods to assure safety and efficacy.
1.3 Although this resin has been used for specific implant applications in the United States, the use of this resin in medical devices should be restricted to non-implant applications until biocompatibility evaluations appropriate for the intended applications are successfully completed.
1.4 The biocompatibility of plastic compounds made up of polyoxymethylene (acetal) resin containing colorants, fillers, processing aids, or other additives as well as polymer blends which contain polyacetal should not be assumed on the basis of resin biocompatibility alone. Their biocompatibility must be established by testing the final (end-use) compositions using evaluation methods appropriate for the intended applications. It should be noted that the types, test levels, and biological effects of extractives yielded by the additives contained in a compound or blend may also have to be evaluated for some end-use applications.
1.5 The values stated in inch-pound units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.8 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D570 Test Method for Water Absorption of Plastics
D638 Test Method for Tensile Properties of Plastics
D695 Test Method for Compressive Properties of Rigid Plastics
D732 Test Method for Shear Strength of Plastics by Punch Tool
D792 Test Methods for Density and Specific Gravity (Relative Density) of Plastics by Displacement
D883 Terminology Relating to Plastics
D1600 Terminology for Abbreviated Terms Relating to Plastics
D1822 Test Method for Tensile-Impact Energy to Break Plastics and Electrical Insulating Materials
D4181 Classification for Acetal (POM) Molding and Extrusion Materials
F748 Practice for Selecting Generic Biological Test Methods for Materials and Devices
ICS Number Code 11.040.30 (Surgical instruments and materials); 83.080.20 (Thermoplastic materials)
UNSPSC Code 13111030(Polyacetal); 42295500(Surgical implants and expanders and extenders and surgical wires and related products)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F1855-00(2019), Standard Specification for Polyoxymethylene (Acetal) for Medical Applications, ASTM International, West Conshohocken, PA, 2019, www.astm.orgBack to Top