| ||Format||Pages||Price|| |
|5||$48.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$48.00||  ADD TO CART|
|Standard + Redline PDF Bundle||10||$58.00||  ADD TO CART|
This specification covers total wrist implants that provide articulation function through radial carpal components. However, this specification does not include implants with ceramic coated or porous coated surfaces, one-piece elastomeric implants, and those used for custom applications. Implants covered by this specification are of the constrained, partially constrained, or unconstrained joint prosthesis type. The implants shall be manufactured from biocompatible materials made of unalloyed titanium, cobalt-28chromium-6molybdenum alloy, wrought cobalt-20chromium-15tungsten-10nickel alloy, wrought titanium-6aluminum-4vanadium extra low interstitial alloy, wrought 35cobalt-35nickel-20chromium-10molybdenum alloy, wrought cobalt-20nickel-20chromium-3.5molybdenum-3.5tungsten-5iron alloy, cobalt-28chromium-6molybdenum alloy, titanium-6aluminum-4vanadium alloy, or wrought cobalt-28chromium-6molybdenum alloy with dimensions in conformity to the requirements specified. Tests for polymeric creep (cold flow), wear, and range of motion of the device as well as in vitro laboratory testing shall be performed and shall conform to the requirements specified. Per service requirement, metallic implants shall undergo fluorescent penetrant inspection, while cast metallic implants shall undergo radiography.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification describes total wrist implants, including solid ceramic implants, used to provide functioning articulation by employing radial and carpal components.
1.2 This specification excludes those implants with ceramic-coated or porous-coated surfaces, one-piece elastomeric implants (with or without grommets), and those devices used for custom applications.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
F67 Specification for Unalloyed Titanium, for Surgical Implant Applications (UNS R50250, UNS R50400, UNS R50550, UNS R50700)
F75 Specification for Cobalt-28 Chromium-6 Molybdenum Alloy Castings and Casting Alloy for Surgical Implants (UNS R30075)
F86 Practice for Surface Preparation and Marking of Metallic Surgical Implants
F90 Specification for Wrought Cobalt-20Chromium-15Tungsten-10Nickel Alloy for Surgical Implant Applications (UNS R30605)
F136 Specification for Wrought Titanium-6Aluminum-4Vanadium ELI (Extra Low Interstitial) Alloy for Surgical Implant Applications (UNS R56401)
F562 Specification for Wrought 35Cobalt-35Nickel-20Chromium-10Molybdenum Alloy for Surgical Implant Applications (UNS R30035)
F563 Specification for Wrought Cobalt-20Nickel-20Chromium-3.5Molybdenum-3.5Tungsten-5Iron Alloy for Surgical Implant Applications (UNS R30563)
F601 Practice for Fluorescent Penetrant Inspection of Metallic Surgical Implants
F603 Specification for High-Purity Dense Aluminum Oxide for Medical Application
F629 Practice for Radiography of Cast Metallic Surgical Implants
F648 Specification for Ultra-High-Molecular-Weight Polyethylene Powder and Fabricated Form for Surgical Implants
F746 Test Method for Pitting or Crevice Corrosion of Metallic Surgical Implant Materials
F748 Practice for Selecting Generic Biological Test Methods for Materials and Devices
F799 Specification for Cobalt-28Chromium-6Molybdenum Alloy Forgings for Surgical Implants (UNS R31537, R31538, R31539)
F981 Practice for Assessment of Compatibility of Biomaterials for Surgical Implants with Respect to Effect of Materials on Muscle and Bone
F983 Practice for Permanent Marking of Orthopaedic Implant Components
F1108 Specification for Titanium-6Aluminum-4Vanadium Alloy Castings for Surgical Implants (UNS R56406)
F1537 Specification for Wrought Cobalt-28Chromium-6Molybdenum Alloys for Surgical Implants (UNS R31537, UNS R31538, and UNS R31539)
ANSI/ASME StandardANSI/ASME B46.1 Surface Texture (Surface Roughness, Waviness, and Lay) Available from American National Standards Institute (ANSI), 25 W. 43rd St., 4th Floor, New York, NY 10036, http://www.ansi.org.
ICS Number Code 11.040.40 (Implants for surgery, prothetics and orthotics)
UNSPSC Code 42295500(Surgical implants and expanders and extenders and surgical wires and related products)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F1357-14, Standard Specification for Articulating Total Wrist Implants, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top