| ||Format||Pages||Price|| |
|5||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$52.00||  ADD TO CART|
Significance and Use
5.1 This test method is intended to measure volatile extractables that may be emitted from a microwave susceptor material during use. It may be a useful procedure to assist in minimizing the amount of volatile extractables either through susceptor design or manufacturing processes.
5.2 Modification of this procedure by utilizing appropriate qualitative GC detection such as a mass spectrometer in place of the flame ionization detector may provide identification of volatile extractables of unknown identity.
1.1 This test method covers complete microwave susceptors.
1.2 This test method covers a procedure for quantitating volatile compounds whose identity has been established and which are evolved when a microwave susceptor sample is tested under simulated use conditions.
1.3 This test method was collaboratively evaluated with a variety of volatile compounds (see statistical evaluation). For compounds other than those evaluated, the analyst should determine the sensitivity and reproducibility of the method by carrying out appropriate spike and recovery studies. The analyst is referred to Practice for guidance.
1.4 For purposes of verifying the identity of or identifying unknown volatile compounds, the analyst is encouraged to incorporate techniques such as gas chromatography/mass spectroscopy, gas chromatography/infrared spectroscopy, or other techniques in conjunction with this test method.
1.5 A sensitivity level of approximately 0.025 μg/in. is achievable for the compounds studied in . Where other compounds are being quantitated and uncertainty exists over method sensitivity, the analyst is referred to Practice for procedures on determining sensitivity of chromatographic methods.
1.6 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use. Specific safety hazards warnings are given in , , and .
1.8 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
TAPPI StandardsTIS 808 Equilibrium relative humidities over saturated salt solutions T402 Standard conditioning and testing atmospheres for paper, board, pulp handsheets, and related products Available from Technical Association of the Pulp and Paper Industry (TAPPI), 15 Technology Parkway South, Norcross, GA 30092, http://www.tappi.org.
E260 Practice for Packed Column Gas Chromatography
F1317 Test Method for Calibration of Microwave Ovens
ICS Number Code 67.250 (Materials and articles in contact with foodstuff and water)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM F1308-98(2018), Standard Test Method for Quantitating Volatile Extractables in Microwave Susceptors Used for Food Products, ASTM International, West Conshohocken, PA, 2018, www.astm.orgBack to Top