| ||Format||Pages||Price|| |
|14||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||14||$54.00||  ADD TO CART|
|Standard + Redline PDF Bundle||28||$65.00||  ADD TO CART|
This practice covers procedures for the verification and classification of extensometer systems, but it is not intended to be a complete purchase specification. The practice is applicable only to instruments that indicate or record values that are proportional to changes in length corresponding to either tensile or compressive strain. Extensometer systems are classified on the basis of the magnitude of their errors. The apparatus for verifying extensometer systems shall provide a means for applying controlled displacements to a simulated specimen and for measuring these displacements accurately. Extensometer systems shall be classified in accordance with the requirements as to maximum error of strain indicated: Class A; Class B-1; Class B-2; Class C; Class D; and Class E. Extensometer systems shall be categorized in three types according to gage length: Type 1; Type 2; and Type 3. A verification procedure for extensometer systems shall be done in accordance with the specified requirements.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This practice covers procedures for the verification and classification of extensometer systems, but it is not intended to be a complete purchase specification. The practice is applicable only to instruments that indicate or record values that are proportional to changes in length corresponding to either tensile or compressive strain. Extensometer systems are classified on the basis of the magnitude of their errors.
1.2 Because strain is a dimensionless quantity, this document can be used for extensometers based on either SI or US customary units of displacement.
Note 1: Bonded resistance strain gauges directly bonded to a specimen cannot be calibrated or verified with the apparatus described in this practice for the verification of extensometers having definite gauge points. (See procedures as described in Test Methods .)
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E6 Terminology Relating to Methods of Mechanical Testing
E21 Test Methods for Elevated Temperature Tension Tests of Metallic Materials
E251 Test Methods for Performance Characteristics of Metallic Bonded Resistance Strain Gages
Other StandardsJCGM 100:2008 Evaluation of measurement data Guide to the expression of uncertainty in measurement
ICS Number Code 19.060 (Mechanical testing)
UNSPSC Code 41114503(Extensometers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E83-16, Standard Practice for Verification and Classification of Extensometer Systems, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top