| ||Format||Pages||Price|| |
|13||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$52.00||  ADD TO CART|
Significance and Use
4.1 The test method described in this standard will ensure that the thermometers listed in Specifications and will indicate temperatures within the maximum scale errors listed, be compatible with the apparatus, and serve the purpose for which they were designed.
4.2 Thermometers that do not pass the visual and dimensional inspection tests may give erroneously high or low temperature readings, or may not fit into existing equipment used in ASTM methods. For accurate temperature measurements the scale readings of the thermometer should be verified as described in this test method.
1.1 This test method covers visual and dimensional inspection and test for scale accuracy to be used in the verification of liquid-in-glass thermometers as specified in Specifications and . However, these procedures may be applied to other liquid-in-glass thermometers.
Note 1: The use of NIST SP250-23 is recommended.
1.2 Warning—Mercury has been designated by EPA and many state agencies as a hazardous material that can cause central nervous system, kidney and liver damage. Mercury, or its vapor, may be hazardous to health and corrosive to materials. Caution should be taken when handling mercury and mercury containing products. See the applicable product Material Safety Data Sheet (MSDS) for details and EPA’s website- http://www.epa.gov/mercury/faq.htm - for additional information. Users should be aware that selling mercury and/or mercury containing products into your state may be prohibited by state law.-
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E1 Specification for ASTM Liquid-in-Glass Thermometers
E344 Terminology Relating to Thermometry and Hydrometry
E2251 Specification for Liquid-in-Glass ASTM Thermometers with Low-Hazard Precision Liquids
ICS Number Code 17.200.20 (Temperature-measuring instruments)
UNSPSC Code 41112239(Glass thermometer)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E77-14e1, Standard Test Method for Inspection and Verification of Thermometers, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top