| ||Format||Pages||Price|| |
|6||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$46.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$55.20||  ADD TO CART|
Significance and Use
4.1 This test method provides for the determination of moisture and residue in liquid chlorine. The concentration levels of these impurities are important factors in many commercial uses of liquid chlorine.
1.1 This test method covers the determination of moisture and residue in liquid chlorine. Lower limits of detection of 10 μg/g based on a 150-mL sample are achievable.
1.2 This test method describes operations and special apparatus for sampling liquid chlorine from cylinders. Sampling other commercial facilities such as tank cars, barges, and storage tanks require special techniques of manipulation in filling sample cylinders for testing. If conditions and facilities are favorable, the analysis apparatus may be connected directly to these larger storage and transportation units and on-the-spot tests may be completed.
1.3 The analytical methods are empirical and are gravimetric. They are applicable to the determination of the residue and moisture content of commercial grades of liquid chlorine (condensed gaseous product).
1.4 Residue is defined as being those substances which remain in the sample flask after sample volatilization under the conditions of the test. Moisture is defined as the volatile substances evolved during volatilization and purging of the sample-residue flask and absorbed on the desiccant contained in the absorption tubes under the conditions of the test. Some of the volatile chlorinated hydrocarbons may also be evolved from the sample residue flask during purge and absorbed, thus contributing to the apparent moisture analysis value.
1.5 This test method may be used for determining the moisture content only of gaseous chlorine if suitable and appropriate modifications are made for sample measurement and calculation.
1.6 Review the current Safety Data Sheet (SDS) for detailed information concerning toxicity, first aid procedures, and safety precautions.
1.7 The values stated in inch-pound units are to be regarded as the standard. No other units of measurement are included in this standard.
1.8 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Specific hazards statements are given in Section .
1.9 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D6809 Guide for Quality Control and Quality Assurance Procedures for Aromatic Hydrocarbons and Related Materials
E180 Practice for Determining the Precision of ASTM Methods for Analysis and Testing of Industrial and Specialty Chemicals
Federal Standards173.304 Charging of Cylinders with Liquefied Compressed Gas 173.314 Requirements for Compressed Gases in Tank Cars 173.315 Compressed Gases in Cargo Tanks and Portable Tank Containers CFR 173 Title 49 Transportation; Shippers General Requirements for Shipments and Packagings, including Sections:
ICS Number Code 71.060.10 (Chemical elements)
UNSPSC Code 12141901(Chlorine Cl)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E410-17a, Standard Test Method for Moisture and Residue in Liquid Chlorine, ASTM International, West Conshohocken, PA, 2017, www.astm.orgBack to Top