| ||Format||Pages||Price|| |
|6||$50.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$50.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$60.00||  ADD TO CART|
Significance and Use
5.1 Information concerning the reaction model aids in the selection of the appropriate method (and test method) for evaluation of kinetic parameters. nth order reaction may be treated by isoconversion methods such as Test Methods and . Autocatalytic reactions are treated by Test Methods .
5.2 This practice may be used in research, forensic analysis, trouble shooting, product evaluation, and hazard potential evaluation.
1.1 This practice describes a procedure for determining the “model” of an exothermic reaction using differential scanning calorimetry. The procedure is typically performed on 1–3 mg specimen sizes over the temperature range from ambient to 600 °C.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E473 Terminology Relating to Thermal Analysis and Rheology
E537 Test Method for Thermal Stability of Chemicals by Differential Scanning Calorimetry
E698 Test Method for Kinetic Parameters for Thermally Unstable Materials Using Differential Scanning Calorimetry and the Flynn/Wall/Ozawa Method
E967 Test Method for Temperature Calibration of Differential Scanning Calorimeters and Differential Thermal Analyzers
E968 Practice for Heat Flow Calibration of Differential Scanning Calorimeters
E1142 Terminology Relating to Thermophysical Properties
E2070 Test Methods for Kinetic Parameters by Differential Scanning Calorimetry Using Isothermal Methods
E2890 Test Method for Kinetic Parameters for Thermally Unstable Materials by Differential Scanning Calorimetry Using the Kissinger Method
ICS Number Code 17.200.10 (Heat. Calorimetry)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E3174-20, Standard Practice for Determination of Kinetic Reaction Model Using Differential Scanning Calorimetry, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top