| ||Format||Pages||Price|| |
|12||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||12||$52.00||  ADD TO CART|
Significance and Use
4.1 Generally, Raman spectra measured using grating-based dispersive or Fourier transform Raman spectrometers have not been corrected for the instrumental response (spectral responsivity of the detection system). Raman spectra obtained with different instruments may show significant variations in the measured relative peak intensities of a sample compound. This is mainly as a result of differences in their wavelength-dependent optical transmission and detector efficiencies. These variations can be particularly large when widely different laser excitation wavelengths are used, but can occur when the same laser excitation is used and spectra of the same compound are compared between instruments. This is illustrated in Fig. 1, which shows the uncorrected luminescence spectrum of SRM 2241, acquired upon four different commercially available Raman spectrometers operating with 785 nm laser excitation. Instrumental response variations can also occur on the same instrument after a component change or service work has been performed. Each spectrometer, due to its unique combination of filters, grating, collection optics and detector response, has a very unique spectral response. The spectrometer dependent spectral response will of course also affect the shape of Raman spectra acquired upon these systems. The shape of this response is not to be construed as either “good or bad” but is the result of design considerations by the spectrometer manufacturer. For instance, as shown in Fig. 1, spectral coverage can vary considerably between spectrometer systems. This is typically a deliberate tradeoff in spectrometer design, where spectral coverage is sacrificed for enhanced spectral resolution.
1.1 This guide is designed to enable the user to correct a Raman spectrometer for its relative spectral-intensity response function using NIST Standard Reference Materials2 in the 224X series (currently SRMs 2241, 2242, 2243, 2244, 2245, 2246), or a calibrated irradiance source. This relative intensity correction procedure will enable the intercomparison of Raman spectra acquired from differing instruments, excitation wavelengths, and laboratories.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E131 Terminology Relating to Molecular Spectroscopy
E1840 Guide for Raman Shift Standards for Spectrometer Calibration
E2529 Guide for Testing the Resolution of a Raman Spectrometer
ANSI StandardZ136.1 Safe Use of Lasers
ICS Number Code 71.040.50 (Physicochemical methods of analysis)
UNSPSC Code 41115403(Spectrometers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2911-13, Standard Guide for Relative Intensity Correction of Raman Spectrometers, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top