| ||Format||Pages||Price|| |
|6||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||12||$54.00||  ADD TO CART|
Significance and Use
5.1 A thermocouple connector, exposed to a temperature difference, contributes to the output of a thermocouple circuit. The output uncertainty allocated to the connector depends on the connector design and temperature gradient.
1.1 This standard describes a thermal emf test method for base-metal thermocouple connectors including Types E, J, K, N and T. Standard connectors such as found in Specifications E1129/E1129M and E1684 as well as non-standard connector configurations and connector components can be evaluated using this method.
1.2 The measured emf is reported as an equivalent temperature deviation or error relative to a reference thermocouple of the same type. This method can be used to verify deviations introduced by the connector greater than or equal to 1°C.
1.3 The connector is tested with thermocouple contacts axially aligned with a temperature gradient using a specified thermal boundary condition. The actual temperature difference developed across the connector and corresponding error will depend on the connector design.
1.4 Connector contacts are often fabricated from raw materials having temperature-emf relationships in accordance with Specification E230. However, verifying Specification E230 tolerances is not within the scope of this method.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E220 Test Method for Calibration of Thermocouples By Comparison Techniques
E230 Specification and Temperature-Electromotive Force (EMF) Tables for Standardized Thermocouples
E344 Terminology Relating to Thermometry and Hydrometry
E563 Practice for Preparation and Use of an Ice-Point Bath as a Reference Temperature
E1129/E1129M Specification for Thermocouple Connectors
E2488 Guide for the Preparation and Evaluation of Liquid Baths Used for Temperature Calibration by Comparison
ICS Number Code 17.200.20 (Temperature-measuring instruments)
UNSPSC Code 41112206(Thermocouples); 39121409(Wire connectors)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2820-13, Standard Test Method for Evaluating Thermal EMF Properties of Base-Metal Thermocouple Connectors, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top