| ||Format||Pages||Price|| |
|3||$42.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$42.00||  ADD TO CART|
Significance and Use
5.1 Refer to Guide E844 for the selection, irradiation, and quality control of neutron dosimeters.
5.2 Refer to Practice E261 for a general discussion of the determination of fast-neutron fluence rate with threshold detectors.
5.6 A second 9.04 h isomer, 58mCo, is formed that decays to 70.86-day 58Co. Loss of 58Co and 58mCo by thermal-neutron burnout will occur in environments5,6 having thermal fluence rates of 3 × 1012 cm−2·s −1 and above. Burnout correction factors, R, are plotted as a function of time for several thermal fluxes in Fig. 1. Tabulated values for a continuous irradiation time are provided in Hogg, et al.6
5.7 Fig. 2 shows a plot of cross section7 versus energy for the fast-neutron reaction 58Ni(n,p) 58Co. This figure is for illustrative purposes only to indicate the range of response of the 58Ni(n,p) reaction. Refer to Guide E1018 for descriptions of recommended tabulated dosimetry cross sections.
1.2 This activation reaction is useful for measuring neutrons with energies above approximately 2.1 MeV and for irradiation times up to about 200 days in the absence of high thermal neutron fluence rates (for longer irradiations, see Practice E261).
1.4 Detailed procedures for other fast-neutron detectors are referenced in Practice E261.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E170 Terminology Relating to Radiation Measurements and Dosimetry
E181 Test Methods for Detector Calibration and Analysis of Radionuclides
E261 Practice for Determining Neutron Fluence, Fluence Rate, and Spectra by Radioactivation Techniques
E844 Guide for Sensor Set Design and Irradiation for Reactor Surveillance, E 706 (IIC)
E944 Guide for Application of Neutron Spectrum Adjustment Methods in Reactor Surveillance, E 706 (IIA)
E1005 Test Method for Application and Analysis of Radiometric Monitors for Reactor Vessel Surveillance, E 706 (IIIA)
E1018 Guide for Application of ASTM Evaluated Cross Section Data File, Matrix E706 (IIB)
ICS Number Code 17.240 (Radiation measurements); 27.120.30 (Fissile materials and nuclear fuel technology)
UNSPSC Code 26142004(Neutron irradiators)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E264-08(2013), Standard Test Method for Measuring Fast-Neutron Reaction Rates by Radioactivation of Nickel, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top