| ||Format||Pages||Price|| |
|13||$58.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$58.00||  ADD TO CART|
|Standard + Redline PDF Bundle||26||$69.00||  ADD TO CART|
Significance and Use
5.1 Users of fire test data often need a quantitative indication of the quality of the data presented in a test report. This quantitative indication is referred to as the “measurement uncertainty”. There are two primary reasons for estimating the uncertainty of fire test results.
5.1.1 ISO/IEC 17025 requires that competent testing and calibration laboratories include uncertainty estimates for the results that are presented in a report.
5.1.2 Fire safety engineers need to know the quality of the input data used in an analysis to determine the uncertainty of the outcome of the analysis.
1.1 This guide covers the evaluation and expression of uncertainty of measurements of fire test methods developed and maintained by ASTM International, based on the approach presented in the GUM. The use in this process of precision data obtained from a round robin is also discussed.
1.2 The guidelines presented in this standard can also be applied to evaluate and express the uncertainty associated with fire test results. However, it may not be possible to quantify the uncertainty of fire test results if some sources of uncertainty cannot be accounted for. This problem is discussed in more detail in .
1.3 Application of this guide is limited to tests that provide quantitative results in engineering units. This includes, for example, methods for measuring the heat release rate of burning specimens based on oxygen consumption calorimetry, such as Test Method .
1.4 This guide does not apply to tests that provide results in the form of indices or binary results (for example, pass/fail). For example, the uncertainty of the Flame Spread Index obtained according to Test Method cannot be determined.
1.5 In some cases additional guidance is required to supplement this standard. For example, the expression of uncertainty of heat release rate measurements at low levels requires additional guidance and uncertainties associated with sampling are not explicitly addressed.
1.6 This fire standard cannot be used to provide quantitative measures.
1.7 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E84 Test Method for Surface Burning Characteristics of Building Materials
E119 Test Methods for Fire Tests of Building Construction and Materials
E176 Terminology of Fire Standards
E230 Specification and Temperature-Electromotive Force (EMF) Tables for Standardized Thermocouples
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E1354 Test Method for Heat and Visible Smoke Release Rates for Materials and Products Using an Oxygen Consumption Calorimeter
ICS Number Code 13.220.40 (Ignitability and burning behaviour of materials and products)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2536-15a, Standard Guide for Assessment of Measurement Uncertainty in Fire Tests, ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top