| ||Format||Pages||Price|| |
|4||$40.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$40.00||  ADD TO CART|
Significance and Use
1.1 This test method describes the calibration or performance conformance for the torque signal generated by commercial or custom-built rheometers. The specific range of the test depends upon the torque range of the rheometer.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E4 Practices for Force Verification of Testing Machines
E473 Terminology Relating to Thermal Analysis and Rheology
E617 Specification for Laboratory Weights and Precision Mass Standards
E1142 Terminology Relating to Thermophysical Properties
ICS Number Code 17.060 (Measurement of volume, mass, density, viscosity)
UNSPSC Code 41112502(Rheometers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2510-07(2013), Standard Test Method for Torque Calibration or Conformance of Rheometers, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top