| ||Format||Pages||Price|| |
|4||$44.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$44.00||  ADD TO CART|
|Standard + Redline PDF Bundle||8||$52.00||  ADD TO CART|
Significance and Use
5.1 The test method calibrates or demonstrates conformity of the torque signal of a rheometer at ambient temperature.
5.2 A calibration factor thus determined may be used to obtain correct torque values.
5.3 This test method may be used in research, development, specification acceptance, and quality control or assurance.
1.1 This test method describes the calibration or performance conformance for the torque signal generated by commercial or custom-built rheometers. The specific range of the test depends upon the torque range of the rheometer.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E4 Practices for Force Verification of Testing Machines
E473 Terminology Relating to Thermal Analysis and Rheology
E617 Specification for Laboratory Weights and Precision Mass Standards
E1142 Terminology Relating to Thermophysical Properties
ICS Number Code 17.060 (Measurement of volume, mass, density, viscosity)
UNSPSC Code 41112502(Rheometers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2510-20, Standard Test Method for Torque Calibration or Conformance of Rheometers, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top