| ||Format||Pages||Price|| |
|4||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$46.00||  ADD TO CART|
Significance and Use
4.1 A key consideration with any radiographic system is its contrast resolution and spatial resolution capability (that is, sensitivity). The degree of obtainable sensitivity with a given system is dependent upon several radiographic parameters such as source energy level, film system class, type and thickness of intensifying screens, and material thickness radiographed. These reference radiographs permit the user to estimate the degree of sensitivity change that may be obtained when these parameters are varied from a specific technique. This standard may also be used in conjunction with Test Method or with Practice or Test Method to provide a basis for developing data for evaluation of a user's specific system. This data may assist a user in determining appropriate parameters for obtaining desired degrees of radiographic system sensitivity.
1.1 This document describes the appearance of a radiographic image where fundamental components of image quality are changed, that is, variables such as whether an X-ray or gamma ray source was used, the characteristics of the radiographic film (gradient, granularity, and developing conditions) and intensifying screens, and specimen thickness (the geometrical configuration of the radiographic set-up and focal spot size both affect image quality but are not considered in the reference radiographs).
1.2 The X-ray film systems used in obtaining the illustrative data were as follows: Very Fine Grain (comparable to class I of Test Method ) and Fine Grain (comparable to class II of Test Method ).
1.3 These reference radiographs consist of four composite illustrations and show how such factors as radiation energy, specimen thickness, and film properties affect the radiographic image. The reference radiograph films are an adjunct to this document and must be purchased separately from ASTM if needed.
1.4 The values stated in inch-pound units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E94 Guide for Radiographic Examination Using Industrial Radiographic Film
E746 Practice for Determining Relative Image Quality Response of Industrial Radiographic Imaging Systems
E1316 Terminology for Nondestructive Examinations
E1735 Practice for Determining Relative Image Quality Response of Industrial Radiographic Imaging Systems from 4 to 25 MeV
E1815 Test Method for Classification of Film Systems for Industrial Radiography
ICS Number Code 19.100 (Non-destructive testing)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E242-15(2020), Standard Reference Radiographs for Appearances of Radiographic Images as Certain Parameters are Changed, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top