| ||Format||Pages||Price|| |
|5||$48.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$48.00||  ADD TO CART|
|Standard + Redline PDF Bundle||10||$58.00||  ADD TO CART|
Significance and Use
4.1 This test method is intended to provide a means for determining the concentration of argon, oxygen, and nitrogen gases in individual sealed insulating glass units, which were intended to be filled with a specific concentration of argon at the time of manufacture.
4.2 The argon, oxygen, and nitrogen are physically separated by gas chromatography and compared to corresponding components separated under similar conditions from a reference standard mixture or mixtures of known composition.
4.3 The composition of the sample is calculated from the chromatogram by comparing the area under the curve of each component with the area under the curve of the corresponding component on the reference standard chromatogram.
4.4 It is essential that the person or persons performing this test are very knowledgeable about the principles and techniques of gas chromatography, operation and calibration of gas chromatographs. More information can be found in Practice E355.
4.5 It takes time for the fill gas to equilibrate in any insulating glass unit. This is particularly important in insulating glass units using a tubular spacer and in units containing interior components such as tubular muntin bars. Performing this test before a unit has equilibrated could result in fill gas concentrations that are measurably different than the actual fill gas concentration.
4.6 This method may be used to determine the initial argon gas concentration achieved by the filling method, or the argon gas concentration in units which have been in service or which have been subjected to durability tests such as those described in Test Methods E773 and E2188.
4.7 This method is not applicable to units filled with mixtures of argon and gases other than air.
4.8 This is a destructive test method in that the edge seal of the insulating glass unit is breached in order to obtain a gas sample for analysis by gas chromatography.
4.9 The argon concentration in the gas fill is part of the information necessary to estimate the thermal performance of the sealed insulating glass unit.
1.1 This test method covers procedures for using gas chromatographs to determine the concentration of argon gas in the space between the panes of sealed insulating glass.
1.2 This test method is not applicable to insulating glass units containing open capillary/breather tubes.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C162 Terminology of Glass and Glass Products
C717 Terminology of Building Seals and Sealants
E177 Practice for Use of the Terms Precision and Bias in ASTM Test Methods
E355 Practice for Gas Chromatography Terms and Relationships
E631 Terminology of Building Constructions
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E773 Test Method for Accelerated Weathering of Sealed Insulating Glass Units
E2188 Test Method for Insulating Glass Unit Performance
ICS Number Code 71.040.50 (Physicochemical methods of analysis); 81.040.01 (Glass in general)
UNSPSC Code 30171710(Insulating glass)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2269-14, Standard Test Method for Determining Argon Concentration in Sealed Insulating Glass Units using Gas Chromatography, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top