| ||Format||Pages||Price|| |
|4||$42.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$42.00||  ADD TO CART|
|Standard + Redline PDF Bundle||8||$50.00||  ADD TO CART|
Significance and Use
5.1 This test method calibrates or demonstrates conformity of a dynamic mechanical analyzer at an isothermal temperature within the range of –100 °C to 300 °C.
5.2 Dynamic mechanical analysis experiments often use temperature ramps. This method does not address the effect of that change in temperature on the storage modulus.
5.3 A calibration factor may be required to obtain corrected storage modulus values.
5.4 This method may be used in research and development, specification acceptance, and quality control or assurance.
1.1 This test method describes the calibration or performance confirmation for the storage modulus scale of a commercial or custom built dynamic mechanical analyzer (DMA) over the temperature range of –100 °C to 300 °C using reference materials in the range of 1 GPa to 200 GPa.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D638 Test Method for Tensile Properties of Plastics
E473 Terminology Relating to Thermal Analysis and Rheology
E698 Test Method for Kinetic Parameters for Thermally Unstable Materials Using Differential Scanning Calorimetry and the Flynn/Wall/Ozawa Method
E1142 Terminology Relating to Thermophysical Properties
E2425 Test Method for Loss Modulus Conformance of Dynamic Mechanical Analyzers
ICS Number Code 17.020 (Metrology and measurement in general)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2254-18, Standard Test Method for Storage Modulus Calibration of Dynamic Mechanical Analyzers, ASTM International, West Conshohocken, PA, 2018, www.astm.orgBack to Top