| ||Format||Pages||Price|| |
|18||$51.00||  ADD TO CART|
|Hardcopy (shipping and handling)||18||$51.00||  ADD TO CART|
|Standard + Redline PDF Bundle||36||$61.20||  ADD TO CART|
This specification covers liquid-in-glass ASTM thermometers using low hazard thermometric liquids. The gas filling above the liquid shall be nitrogen or other suitable inert gas. The filling gas shall be chosen to have very low solubility in the thermometric fluid. The stem shall be made of suitable thermometer tubing and shall have a plain front and enamel back. The bulb shall be made of glass and the following distances between graduations and the bulb, and between graduations and enlargements in the capillary, are minimum limits acceptable. All graduation lines, figures, and letters shall be clearly defined, suitably colored, and permanent. The width and the sharpness of the graduation lines shall be designed in accordance with necessary space between the graduations and the desired accuracy of interpolation. The middle of the graduation line shall be accurately determinable. In addition, the graduation lines shall be straight, of uniform width, and perpendicular to the axis of the thermometer. On partial immersion thermometers an immersion line shall be permanently marked on the front of the thermometer at the distance above the bottom of the bulb as specified. The immersion inscription shall be written in capital letters and abbreviated. The terminal number shall be in full when there are one or more numbered graduations between it and the next full number. The special inscription specified shall be marked on the thermometer in capital letters and Arabic numbers without the use of periods.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 The purpose of this standard is to specify liquid-in-glass ASTM thermometers using low hazard thermometric liquids defined in this standard.
1.2 This standard specifies liquid-in-glass thermometers graduated in degrees Celsius or degrees Fahrenheit that are frequently identified and used in methods under the jurisdiction of the various technical committees within ASTM. The current approved thermometers are listed in .
1.3 The technical requirements for the thermometric liquids used in the thermometers in are specified in . Tests for conformity to the technical requirements are also found in .
Note 1: It has been found by experience that ASTM Thermometers, although developed in general for specific tests, may also be found suitable for other applications, thus precluding the need for new thermometer specifications differing in only minor features. However, it is suggested that technical committees contact E20.05 before choosing a currently designated thermometer for a new method to be sure the thermometer will be suitable for the intended application.
1.4 For full rationale, see .
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory requirements prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E1 Specification for ASTM Liquid-in-Glass Thermometers
E77 Test Method for Inspection and Verification of Thermometers
E344 Terminology Relating to Thermometry and Hydrometry
E563 Practice for Preparation and Use of an Ice-Point Bath as a Reference Temperature
ICS Number Code 17.200.20 (Temperature-measuring instruments)
UNSPSC Code 41112200(Temperature and heat measuring instruments)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2251-14, Standard Specification for Liquid-in-Glass ASTM Thermometers with Low-Hazard Precision Liquids, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top