| ||Format||Pages||Price|| |
|5||$45.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$45.00||  ADD TO CART|
|Standard + Redline PDF Bundle||10||$54.00||  ADD TO CART|
Significance and Use
4.1 Knowledge of the ultraviolet transmittance of monoethylene glycol is required to establish whether the product meets the requirements of its quality specifications.
4.2 Dissolved oxygen in organic solvents, such as MEG, forms complexes that shift the solvent absorption from the vacuum ultraviolet range into the measurable UV range (near 190 to 250 nm). Monoethylene glycol has a UV absorption peak at 180 nm. For MEG-oxygen complexes, this peak is shifted to a longer wavelength, thus increasing the absorbability at 220 nm.
4.2.1 However, this effect is not observed in water. There is no significant measurable effect due to dissolved oxygen in water that would require nitrogen sparging prior to using for collection of the reference spectrum.
4.2.2 Nitrogen sparging and re-measurement of suspect or borderline glycol samples at 220 nm can be used as a tool to rule out or confirm the presence of UV affecting contaminants other than oxygen.
1.1 This test method covers a procedure for the determination of the transmittance of monoethylene glycol (1,2-ethanediol; MEG) at wavelengths in the region 220 to 350 nm. The results provide a measure of the purity of the sample with respect to ultraviolet absorbing compounds.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific hazard statements, see Section .
1.4 Review the current Safety Data Sheets (SDS) for detailed information concerning toxicity, first aid procedures, and safety precautions for all materials used in this test method.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
D6299 Practice for Applying Statistical Quality Assurance and Control Charting Techniques to Evaluate Analytical Measurement System Performance
E131 Terminology Relating to Molecular Spectroscopy
E169 Practices for General Techniques of Ultraviolet-Visible Quantitative Analysis
E180 Practice for Determining the Precision of ASTM Methods for Analysis and Testing of Industrial and Specialty Chemicals
E275 Practice for Describing and Measuring Performance of Ultraviolet and Visible Spectrophotometers
Other DocumentManufacturer’s Instruction Manual of Spectr
ICS Number Code 71.040.50 (Physicochemical methods of analysis)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2193-16, Standard Test Method for Ultraviolet Transmittance of Monoethylene Glycol (using Ultraviolet Spectrophotometry), ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top