| ||Format||Pages||Price|| |
|3||$41.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$41.00||  ADD TO CART|
|Standard + Redline PDF Bundle||6||$49.20||  ADD TO CART|
Significance and Use
4.1 This practice is suited ideally for screening samples for the presence, relative concentration, and potential class of ignitable liquid residues in fire debris.
4.2 This is a very sensitive separation procedure, capable of isolating small quantities of ignitable liquid residues from a sample, that is, a 0.1 μL spike of gasoline on a cellulose wipe inside of a 1-gal can is detectable.
4.3 Actual recovery will vary, depending on several factors, including adsorption temperature, container size, competition from the sample matrix, ignitable liquid class and relative ignitable liquid concentration.
4.4 Because this separation takes place in a closed container, the sample remains in approximately the same condition in which it was submitted. Repeat and interlaboratory analyses, therefore, may be possible. Since the extraction is nonexhaustive, the technique permits reanalysis of samples.
4.5 This practice is intended for use in conjunction with other extraction techniques described in Practices , , , and .
4.6 The extract is consumed in the analysis. If a more permanent extract is desired, one of the separation practices described in Practices , , or should be used.
1.1 This practice describes the procedure for removing small quantities of ignitable liquid residues from samples of fire debris. An adsorbent material is used to extract the residue from the static headspace above the sample. Then, analytes are thermally desorbed in the injection port of the gas chromatograph (GC).
1.2 This practice is best suited for screening fire debris samples to assess relative ignitable liquid concentration and for extracting ignitable liquid from aqueous samples.
1.3 This practice is suitable for extracting ignitable liquid residues when a high level of sensitivity is required due to a very low concentration of ignitable liquid residues in the sample.
1.3.1 Unlike other methods of separation and concentration, this method recovers a minimal amount of the ignitable residues present in the evidence, leaving residues that are suitable for subsequent resampling.
1.4 Alternate separation and concentration procedures are listed in Section .
1.5 This standard cannot replace knowledge, skill, or ability acquired through appropriate education, training, and experience and should be used in conjunction with sound professional judgment.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E1386 Practice for Separation of Ignitable Liquid Residues from Fire Debris Samples by Solvent Extraction
E1388 Practice for Sampling of Headspace Vapors from Fire Debris Samples
E1412 Practice for Separation of Ignitable Liquid Residues from Fire Debris Samples by Passive Headspace Concentration With Activated Charcoal
E1413 Practice for Separation of Ignitable Liquid Residues from Fire Debris Samples by Dynamic Headspace Concentration
E1492 Practice for Receiving, Documenting, Storing, and Retrieving Evidence in a Forensic Science Laboratory
E1618 Test Method for Ignitable Liquid Residues in Extracts from Fire Debris Samples by Gas Chromatography-Mass Spectrometry
ICS Number Code 27.060.10 (Liquid and solid fuel burners)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E2154-15a, Standard Practice for Separation and Concentration of Ignitable Liquid Residues from Fire Debris Samples by Passive Headspace Concentration with Solid Phase Microextraction (SPME), ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top