| ||Format||Pages||Price|| |
|5||$48.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$48.00||  ADD TO CART|
Significance and Use
4.1 This test method provides a test for determining the relative image quality response of radiographic film when exposed to 4 to 25 MeV X rays as any single component of the total X-ray system (for example, screens) is varied. By holding the technique parameters (except exposure time) and processing parameters constant, the image quality response of radiographic film may be evaluated on a relative basis.
4.2 Alternately, this test method provides a test for measuring the image quality of the X-ray system or any component of the system.
1.1 This test method covers determination of the relative image quality response of industrial radiographic film when exposed to X-radiation sources having photon energies from 4 to 25 MeV. Evaluation of the film is based on the visibility of holes in a special image quality indicator (IQI). Since results for a given film type may vary, depending on the particular processing system and processing conditions used, it is essential to state the exposure parameters and achieved density, processing chemistry, processing cycle, and processing temperature. For the purposes of this test method, it is assumed that all components of the X-ray system are operating properly and are capable of producing a given image quality. This test method is not intended to be used for films exposed with Cobalt 60 sources or X-ray sources below 4 MeV.
1.2 The values stated in either SI or inch-pound units are to be regarded as the standard. The values given in parentheses are for information only.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E999 Guide for Controlling the Quality of Industrial Radiographic Film Processing
E1025 Practice for Design, Manufacture, and Material Grouping Classification of Hole-Type Image Quality Indicators (IQI) Used for Radiology
E1079 Practice for Calibration of Transmission Densitometers
E1316 Terminology for Nondestructive Examinations
E1815 Test Method for Classification of Film Systems for Industrial Radiography
ICS Number Code 37.040.20 (Photographic paper, film and plates. Cartridges)
UNSPSC Code 26142304(Radiographic equipment)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E1735-07(2014), Standard Test Method for Determining Relative Image Quality of Industrial Radiographic Film Exposed to X-Radiation from 4 to 25 MeV, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top