| ||Format||Pages||Price|| |
|4||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$46.00||  ADD TO CART|
Significance and Use
5.1 The purpose of this test method is to provide data on liquid drop-size characteristics for sprays, as indicated by optical nonimaging light-scattering instruments. The results obtained generally will be statistical in nature. The number of variables concerned in the production of liquid spray, together with the variety of optical, electronic, and sampling systems used in different instruments, may contribute to variations in the test results. Care must be exercised, therefore, when attempting to compare data from samples obtained by different means.
1.1 The purpose of this test method is to obtain data which characterize the sizes of liquid particles or drops such as are produced by a spray nozzle or similar device under specified conditions using a specified liquid. The drops will generally be in the size range from 5-μm to the order of 1 000-μm diameter; they will occur in sprays which may be as small as a few cubic centimetres or as large as several cubic metres. Typically the number density of the particles can vary significantly from one point to another.
1.2 This test method is intended primarily for use in standardizing measurements of the performance of sprayproducing devices. It is limited to those techniques and instruments that operate by passing a beam of light through the spray and analyzing the light scattered by the droplets to derive size information. Such techniques do not produce images of individual drops, and therefore, are known as “optical (nonimaging) instruments.”
1.3 The measurements made, when referred to the entire spray being sampled, may be flux sensitive or spatial, as defined in Practice , depending on the techniques used with a particular instrument.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.5 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E177 Practice for Use of the Terms Precision and Bias in ASTM Test Methods
E456 Terminology Relating to Quality and Statistics
E799 Practice for Determining Data Criteria and Processing for Liquid Drop Size Analysis
E1088 Definitions of Terms Relating to Atomizing Devices
E1296 Terminology for Liquid Particle Statistics
E1620 Terminology Relating to Liquid Particles and Atomization
NFPA StandardsNFPA 30 Flammable and Combustible Liquids Code Available from National Fire Protection Association (NFPA), 1 Batterymarch Park, Quincy, MA 02169-7471, http://www.nfpa.org.
ICS Number Code 17.180.30 (Optical measuring instruments); 55.130 (Aerosol containers)
UNSPSC Code 23153143(Spray nozzle)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E1260-03(2020), Standard Test Method for Determining Liquid Drop Size Characteristics in a Spray Using Optical Nonimaging Light-Scattering Instruments, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top