| ||Format||Pages||Price|| |
|17||$58.00||  ADD TO CART|
|Hardcopy (shipping and handling)||17||$58.00||  ADD TO CART|
|Standard + Redline PDF Bundle||34||$69.00||  ADD TO CART|
Significance and Use
5.1 One of the factors affecting the quality of radiologic images is the geometric unsharpness. The degree of geometric unsharpness is dependent on the focal spot size of the radiation source, the distance between the source and the object to be radiographed, and the distance between the object to be radiographed and the detector (imaging plate, Digital Detector Array (DDA) or film). This test method allows the user to determine the effective focal size of the X-ray source. This result may then be used to establish source to object and object to detector distances appropriate for maintaining the desired degree of geometric unsharpness or maximum magnification for a given radiographic imaging application, or both. Some ASTM standards require this value for calculation of a required magnification, for example, Practices , , and .
1.1 The image quality and the resolution of X-ray images are especially sensitive to the characteristics of the focal spot. The imaging qualities of the focal spot are based on its two dimensional intensity distribution as seen from the detector plane.
1.2 This test method provides instructions for determining the effective size (dimensions) of standard and mini focal spots of industrial X-ray tubes for focal spot dimensions from 100 μm up to several mm of X-ray sources up to 600 kV tube voltage. Smaller focal spots down to 50 µm could be evaluated with less precision. This determination is based on the measurement of an image of a focal spot that has been radiographically recorded with a “pinhole” technique. An alternative method with a plaque hole IQI may be found in the Annex A, which covers the same focal spot sizes.
1.3 Smaller focal spots should be measured using Test Method using the projection of an edge.
1.4 This test method may also be used to determine the change in focal spot size that may have occurred due to tube age, tube overloading, and the like. This would entail the production of a focal spot radiograph (with the pinhole method) and an evaluation of the resultant image for pitting, cracking, and the like.
1.5 Units—Values stated in SI units are to be regarded as the standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
E999 Guide for Controlling the Quality of Industrial Radiographic Film Processing
E1000 Guide for Radioscopy
E1025 Practice for Design, Manufacture, and Material Grouping Classification of Hole-Type Image Quality Indicators (IQI) Used for Radiography
E1255 Practice for Radioscopy
E1742 Practice for Radiographic Examination
E1815 Test Method for Classification of Film Systems for Industrial Radiography
E2002 Practice for Determining Total Image Unsharpness and Basic Spatial Resolution in Radiography and Radioscopy
E2033 Practice for Radiographic Examination Using Computed Radiography (Photostimulable Luminescence Method)
E2339 Practice for Digital Imaging and Communication in Nondestructive Evaluation (DICONDE)
E2698 Practice for Radiographic Examination Using Digital Detector Arrays
E2736 Guide for Digital Detector Array Radiography
E2903 Test Method for Measurement of the Effective Focal Spot Size of Mini and Micro Focus X-ray Tubes
European StandardsEN 12543-2 Non-destructive testingCharacteristics of focal spots in industrial X-ray systems for use in non-destructive testingPart 2: Pinhole camera radiographic method EN 12543-5
ICS Number Code 19.100 (Non-destructive testing)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E1165-20, Standard Test Method for Measurement of Focal Spots of Industrial X-Ray Tubes by Pinhole Imaging, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top