| ||Format||Pages||Price|| |
|5||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$52.00||  ADD TO CART|
This specification covers sampling and testing of laboratory glassware items for use until they are no longer considered functional for the intended purpose, and does not include disposable (single use) items. Four types of reusable laboratory glassware are covered, namely: blown and pressed glassware, nongraduated; blown and pressed glassware, graduated; tubular glassware, nongraduated; and tubular glassware, graduated. The product shall be designed to meet the specified test performance requirements. Alternate testing methods, inspection levels, and sample sizes may be implemented. Sampling shall be conducted in accordance with the specified inspection and examination levels and acceptance quality levels for major defects, minor defects, and dimensional examination. Where applicable, distilled water and reagent grade chemicals shall be used throughout the tests. Inspection shall be conducted in accordance with applicable section and individual specifications. The basic testing criteria for the following tests are detailed: (1) leakage test, (2) permanency of marking by alkali and acid tests, (3) capacity fill test, (4) autoclaving, (5) pour test, and (6) centrifugal test.
This abstract is a brief summary of the referenced standard. It is informational only and not an official part of the standard; the full text of the standard itself must be referred to for its use and application. ASTM does not give any warranty express or implied or make any representation that the contents of this abstract are accurate, complete or up to date.
1.1 This specification covers laboratory items for use until they are no longer considered functional for the intended purpose. It is written specifically for reusable items and is not to be confused with disposable (single use) items that are described in other standards.
1.2 This specification covers the sampling inspection and basic testing criteria for the following four types of laboratory glassware: blown and pressed glassware, nongraduated; blown and pressed glassware, graduated; tubular glassware, nongraduated; and tubular glassware, graduated.
1.3 This specification is intended to be used in conjunction with a standard specification for a specific laboratory glassware product.
1.4 The following precautionary statement pertains only to the test method portion, Section , of this specification. This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.5 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C162 Terminology of Glass and Glass Products
E542 Practice for Calibration of Laboratory Volumetric Apparatus
E671 Specification for Maximum Permissible Thermal Residual Stress in Annealed Glass Laboratory Apparatus
ANSI/ASQ StandardANSI/ASQ Z1.4
ICS Number Code 71.040.20 (Laboratory ware and related apparatus)
UNSPSC Code 41104918(Laboratory glass filters)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM E1157-87(2018)e1, Standard Specification for Sampling and Testing of Reusable Laboratory Glassware, ASTM International, West Conshohocken, PA, 2011, www.astm.orgBack to Top