| ||Format||Pages||Price|| |
|5||$50.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$50.00||  ADD TO CART|
Significance and Use
5.1 This test method is applicable for determining the level of iron contamination in lubricating oils from sources other than physical wear of the machinery or equipment, such as corrosion. It is particularly suited for oils from engines running on sulphur containing fuels where acidic combustion by-products can enter the lubrication system and cause corrosive damage. The test can be performed at the point of use as it does not require extensive laboratory equipment allowing timely intervention if necessary.
1.1 This method covers a procedure for determining the concentration of compounded forms of iron in in-service fluids. It is applicable over an iron concentration range of 20 mg/kg to 800 mg/kg (ppm by mass hereafter). The test is not sensitive to metallic iron and thus provides a means of discrimination between the two forms of iron. It is suitable for monitoring the condition of equipment where iron contamination in the lubricating oil or functional fluid may not originate solely from physical wear, for example from corrosion or other forms of undesirable chemical processes within the equipment.
1.2 Units—The values stated in SI units are to be regarded as standard. Concentrations are reported as parts per million by mass (ppm) equivalent to iron contents expressed in mg per kg of oil sample (mg/kg).
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products
D6300 Practice for Determination of Precision and Bias Data for Use in Test Methods for Petroleum Products and Lubricants
E177 Practice for Use of the Terms Precision and Bias in ASTM Test Methods
ICS Number Code 75.100 (Lubricants, industrial oils and related products)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D8304-19, Standard Test Method for Determination of Compounded Forms of Iron in In-service Fluids, ASTM International, West Conshohocken, PA, 2019, www.astm.orgBack to Top