| ||Format||Pages||Price|| |
|13||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$54.00||  ADD TO CART|
Significance and Use
5.1 Appropriate application of this practice should result in a WQE achievable by the laboratory in applying the tested method/matrix/analyte combination to routine sample analysis. That is, a laboratory should be capable of measuring concentrations greater than WQEZ %, with the associated RSD equal to Z % or less.
5.3 The WQE procedure should be used to establish the within-laboratory quantitation capability for any application of a method in the laboratory where quantitation is important to data use. The intent of the WQE is not to impose reporting limits. The intent is to provide a reliable procedure for establishing the quantitative characteristics of the method (as implemented in the laboratory for the matrix and analyte) and thus to provide the laboratory with reliable information characterizing the uncertainty in any data produced. Then the laboratory may make informed decisions about censoring data and has the information necessary for providing reliable estimates of uncertainty with reported data.
1.1 This practice establishes a uniform standard for computing the within-laboratory quantitation estimate associated with Z % relative standard deviation (referred to herein as WQEZ %), and provides guidance concerning the appropriate use and application.
1.2 WQEZ % is computed to be the lowest concentration for which a single measurement from the laboratory will have an estimated Z % relative standard deviation (Z % RSD, based on within-laboratory standard deviation), where Z is typically an integer multiple of 10, such as 10, 20, or 30. Z can be less than 10 but not more than 30. The WQE10 % is consistent with the quantitation approaches of Currie (1)2 and Oppenheimer, et al (2).
1.3 The fundamental assumption of the WQE is that the media tested, the concentrations tested, and the protocol followed in the developing the study data provide a representative and fair evaluation of the scope and applicability of the test method, as written. Properly applied, the WQE procedure ensures that the WQE value has the following properties:
1.3.1 Routinely Achievable WQE Value—The laboratory should be able to attain the WQE in routine analyses, using the laboratory‘s standard measurement system(s), at reasonable cost. This property is needed for a quantitation limit to be feasible in practical situations. Representative data must be used in the calculation of the WQE.
1.3.2 Accounting for Routine Sources of Error—The WQE should realistically include sources of bias and variation that are common to the measurement process and the measured materials. These sources include, but are not limited to intrinsic instrument noise, some typical amount of carryover error, bottling, preservation, sample handling and storage, analysts, sample preparation, instruments, and matrix.
1.3.3 Avoidable Sources of Error Excluded—The WQE should realistically exclude avoidable sources of bias and variation (that is, those sources that can reasonably be avoided in routine sample measurements). Avoidable sources would include, but are not limited to, modifications to the sample, modifications to the measurement procedure, modifications to the measurement equipment of the validated method, and gross and easily discernible transcription errors (provided there was a way to detect and either correct or eliminate these errors in routine processing of samples).
1.4 The WQE applies to measurement methods for which instrument calibration error is minor relative to other sources, because this practice does not model or account for instrument calibration error, as is true of quantiation estimates in general. Therefore, the WQE procedure is appropriate when the dominant source of variation is not instrument calibration, but is perhaps one or more of the following:
1.5 Data Quality Objectives—For a given method, one typically would compute the lowest % RSD possible for any given data set. Thus, if possible, WQE10 % would be computed. If the data indicated that the method was too noisy, one might have to compute instead WQE20 %, or possibly WQE30 %. In any case, a WQE with a higher % RSD level (such as WQE50 %) would not be considered, though a WQE with RSD <10 % (such as WQE1 %) would be acceptable. The appropriate level of % RSD is based on the data-quality objective(s) for a particular use or uses. This practice allows for calculation of WQEs with user selected % RSDs less than 30 %.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1129 Terminology Relating to Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D6091 Practice for 99 %/95 % Interlaboratory Detection Estimate (IDE) for Analytical Methods with Negligible Calibration Error
D6512 Practice for Interlaboratory Quantitation Estimate
D7510 Practice for Performing Detection and Quantitation Estimation and Data Assessment Utilizing DQCALC Software, based on ASTM Practices D6091 and D6512 of Committee D19 on Water
E1763 Guide for Interpretation and Use of Results from Interlaboratory Testing of Chemical Analysis Methods
ICS Number Code 17.020 (Metrology and measurement in general)
UNSPSC Code 81141500(Quality control)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D7783-13, Standard Practice for Determination of the 99 %/95 % Critical Level (WCL) and a Reliable Detection Estimate (WDE) Based on Within-laboratory Data, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top