| ||Format||Pages||Price|| |
|4||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$46.00||  ADD TO CART|
Significance and Use
5.1 The purpose of this practice is to provide standardization in the value setting of calibration cotton standards to be used for cotton classification instrument calibration.
5.2 The practice provides procedures for the value setting for calibration cotton standards. These procedures are for use by cotton fiber testing organizations that establish standard values on candidate bales that will be utilized as calibration cotton standards for cotton classification instruments on the level of the benchmark cottons.
Note 1: Benchmark cottons are of high value and of limited supply. Long term accuracy and stability of the internationally recognized USDA cotton standards are dependent on long term maintenance of the benchmark cottons. Therefore, distribution of the benchmark cottons will be tightly controlled by the USDA.
1.1 This practice provides instructions for the development of calibration cotton standards by establishing cotton fiber values for micronaire, length, uniformity index, and strength. The purpose of calibration cotton standards is for calibration of cotton classification instruments to the level of the internationally recognized United States Department of Agriculture (USDA) Benchmark Reference Cotton Standards. The USDA has established calibration cotton standards for domestic upland and Extra Long Staple (ELS) cottons and this practice provides an instruction for other geographical regions to establish their own calibration cotton standards for cotton classification instruments.
1.2 This practice shall be used for the establishment of calibration cottons for the testing of Upland and ELS cottons.
1.3 Only saw ginned cottons shall be used in this practice. Roller gin cottons are not valid for use due to higher measurement variability.
1.4 The calibration cotton established by this practice is not valid for use as a micronaire calibration standard. The micronaire standard value is not for the instrument calibration of the micronaire module; it is utilized by the instrument for the correction of the strength measurement during the calibration routine.
1.5 The values stated in inch-pound are to be regarded as the standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D123 Terminology Relating to Textiles
D7139 Terminology for Cotton Fibers
D7410 Practice for Qualification of Cotton Classification Instruments for Cotton Marketing
ICS Number Code 59.060.10 (Natural fibres)
UNSPSC Code 11151507(Cotton fibers)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D7642-12(2020), Standard Practice for Establishment of Calibration Cottons for Cotton Classification Instruments, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top