| ||Format||Pages||Price|| |
|19||$56.00||  ADD TO CART|
|Hardcopy (shipping and handling)||19||$56.00||  ADD TO CART|
|Standard + Redline PDF Bundle||38||$67.00||  ADD TO CART|
Significance and Use
5.1 The boiling range distribution of medium and heavy petroleum distillate fractions provides an insight into the composition of feed stocks and products related to petroleum refining processes (for example, hydrocracking, hydrotreating, visbreaking, or deasphalting). The gas chromatographic simulation of this determination can be used to replace conventional distillation methods for control of refining operations. This test method can be used for product specification testing with the mutual agreement of interested parties.
5.2 This test method extends the scope of boiling range determination by gas chromatography to include distillates (IBP > 100 °C) and heavy petroleum distillate fractions beyond the scope of Test Method (538 °C).
5.3 Boiling range distributions obtained by this test method have not been analyzed for correlation to those obtained by low efficiency distillation, such as with Test Method or . This test method does not claim agreement between these physical distillations and simulated distillation. Efforts to resolve this question will continue. When successful resolutions of the questions are determined, this test method will be revised accordingly.
1.1 This test method covers the determination of the boiling range distribution of petroleum products by capillary gas chromatography using flame ionization detection. This standard test method has been developed through the harmonization of two test methods, Test Method and IP 480. As both of these methods cover the same scope and include very similar operating conditions, it was agreed that a single standard method would benefit the global simulated distillation community.
1.2 This test method is not applicable for the analysis of petroleum or petroleum products containing low molecular weight components (for example naphthas, reformates, gasolines, diesel). Components containing hetero atoms (for example alcohols, ethers, acids, or esters) or residue are not to be analyzed by this test method. See Test Methods , , or for possible applicability to analysis of these types of materials. This method is also not suitable for samples that will not elute completely from the gas chromatographic column, leaving residues. For such samples as crude oils and residues, see Test Methods and .
1.3 This test method is applicable to distillates with initial boiling points above 100 ºC and final boiling points below 735 ºC (carbon 110); for example, distillates (IBP > 100 °C), base oils and lubricating base stocks.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D86 Test Method for Distillation of Petroleum Products at Atmospheric Pressure
D1160 Test Method for Distillation of Petroleum Products at Reduced Pressure
D2887 Test Method for Boiling Range Distribution of Petroleum Fractions by Gas Chromatography
D5307 Test Method for Determination of Boiling Range Distribution of Crude Petroleum by Gas Chromatography
D6352 Test Method for Boiling Range Distribution of Petroleum Distillates in Boiling Range from 174 C to 700 C by Gas Chromatography
D7096 Test Method for Determination of the Boiling Range Distribution of Gasoline by Wide-Bore Capillary Gas Chromatography
D7169 Test Method for Boiling Point Distribution of Samples with Residues Such as Crude Oils and Atmospheric and Vacuum Residues by High Temperature Gas Chromatography
D7213 Test Method for Boiling Range Distribution of Petroleum Distillates in the Boiling Range from 100 C to 615 C by Gas Chromatography
E355 Practice for Gas Chromatography Terms and Relationships
E594 Practice for Testing Flame Ionization Detectors Used in Gas or Supercritical Fluid Chromatography
E1510 Practice for Installing Fused Silica Open Tubular Capillary Columns in Gas Chromatographs
ISO StandardISO 3170 Petroleum Liquids Manual Sampling
ICS Number Code 75.100 (Lubricants, industrial oils and related products)
UNSPSC Code 15120000(Lubricants and oils and greases and anti corrosives)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D7500-15, Standard Test Method for Determination of Boiling Range Distribution of Distillates and Lubricating Base Oils—in Boiling Range from 100 °C to 735 °C by Gas Chromatography, ASTM International, West Conshohocken, PA, 2015, www.astm.orgBack to Top