| ||Format||Pages||Price|| |
|13||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$54.00||  ADD TO CART|
|Standard + Redline PDF Bundle||26||$65.00||  ADD TO CART|
Significance and Use
5.1 This test method is intended for the measurement of gross alpha- and beta-activity concentrations in the analyses of environmental and drinking waters. For samples submitted to satisfy regulatory or permit requirements, the submitter should assure that this or any other method used is acceptable to the regulator or permit issuer.
5.2 This test method is also applicable to the direct analysis of gross alpha- and beta-activity concentrations in water when low detection limits are not required. Direct analysis provides a rapid method for determination of gross alpha- and beta-activity concentrations when low detection limits are not required.
5.3 This test method is not capable of discriminating among alpha emitting radionuclides or among beta emitting radionuclides. Those intending to identify and quantify specific radionuclides should use test methods specific to the radionuclides of interest.
5.4 This test method may not be cited as a method for the determination of gross alpha- or beta-activity concentrations in a solid/soil matrix or the acid digestate of the same. The use of this test method for such applications brings the potential for serious bias and incomparability of results dependent on the matrix constituents, manner of sample preparation or treatment, or both.
1.1 This test method covers the measurement of gross alpha- and beta- activity concentrations in a homogeneous water sample. It is applicable to alpha emitters with activity concentration levels above 0.11 Bq/L (3 pCi/L) and beta emitters with activity concentration levels above 0.15 Bq/L (4 pCi/L). This test method is not applicable to samples containing radionuclides that are volatile under conditions of the analysis.
1.2 This test method may also be used for the direct measurement of gross alpha- and beta- activity concentrations in homogeneous water samples with alpha emitter activity concentration levels above 1.8 Bq/L (50 pCi/L) and beta emitter activity concentration levels above 3.7 Bq/L (100 pCi/L).
1.3 This test method was tested using single-operator tests., A collaborative study following the U.S. EPA “Protocol for the Evaluation of Alternate Test Procedures for Analyzing Radioactive Contaminants in Drinking Water” was performed. The results of this study are on file at ASTM Headquarters.
1.4 The values stated in SI units are to be regarded as standard. The values given in parentheses are mathematical conversions to pCi/L that are provided for information only and are not considered standard. An exception is noted in Section .
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1125 Test Methods for Electrical Conductivity and Resistivity of Water
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D1890 Test Method for Beta Particle Radioactivity of Water
D1943 Test Method for Alpha Particle Radioactivity of Water
D3370 Practices for Sampling Water from Closed Conduits
D3648 Practices for the Measurement of Radioactivity
D3856 Guide for Management Systems in Laboratories Engaged in Analysis of Water
D4448 Guide for Sampling Ground-Water Monitoring Wells
D5847 Practice for Writing Quality Control Specifications for Standard Test Methods for Water Analysis
D6001 Guide for Direct-Push Groundwater Sampling for Environmental Site Characterization
D7902 Terminology for Radiochemical Analyses
E177 Practice for Use of the Terms Precision and Bias in ASTM Test Methods
E691 Practice for Conducting an Interlaboratory Study to Determine the Precision of a Test Method
ICS Number Code 17.240 (Radiation measurements)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D7283-17, Standard Test Method for Alpha and Beta Activity in Water By Liquid Scintillation Counting, ASTM International, West Conshohocken, PA, 2017, www.astm.orgBack to Top