| ||Format||Pages||Price|| |
|8||$50.00||  ADD TO CART|
|Hardcopy (shipping and handling)||8||$50.00||  ADD TO CART|
|Standard + Redline PDF Bundle||16||$60.00||  ADD TO CART|
Significance and Use
4.1 Compacted asphalt mixture specimens molded by this procedure are used for various physical tests such as stability, flow, indirect tensile strength, fatigue, creep, and modulus. Density and void analysis are also conducted on specimens for mixture design and evaluation of field compaction.
Note 3: Uncompacted mixtures are used for determination of theoretical maximum specific gravity.
Note 4: The quality of the results produced by this practice are dependent on the competence of the personnel performing the procedure and the capability, calibration, and maintenance of the equipment used. Agencies that meet the criteria of Specification are generally considered capable of competent and objective testing, sampling, inspection, etc. Users of this practice are cautioned that compliance with Specification alone does not completely ensure reliable results. Reliable results depend on many factors; following the suggestions of Specification or some similar acceptable guideline provides a means of evaluating and controlling some of those factors.
1.1 This practice covers preparation and compaction of 4 in. (101.6 mm) diameter by nominal 2.5 in. (63.5 mm) high cylindrical asphalt paving mixture specimens. This practice is intended for use with laboratory and plant-produced asphalt mixtures with aggregate up to 1 in. (25.4 mm) maximum size and for recompaction of asphalt paving mixture samples.
Note 1: Historically, 35, 50, and 75 blows per face has been practiced for this test.
Note 2: Manufacturers do not recommend applying an excessive number of blows per face of a sample. This practice has been known to cause fatigue and damage to machine components.
1.2 There are three types of Marshall compaction apparatus in use. The following types of hammer arrangements are included in this practice:
1.2.1 Manually held hammer handle (Type 1) is attached to a flat compaction foot through a spring-loaded swivel and is hand operated (see ) (original standard developed by the United States Army Corps of Engineers).
1.2.2 Hammer handle restrained laterally (fixed) but not vertically (Type 2), attached to a flat compaction foot through a spring-loaded swivel and is either mechanically or hand operated (see ). There may or may not be a constant surcharge on top of the hammer handle. Mechanical hammers (Type 2) are available that operate at a range of 55 to 68 blows per minute.
1.2.3 Hammer handle restrained laterally (fixed) with or without constant surcharge on top of hammer, is attached to a slanted compaction foot on a rotating mold base (Type 3), and is mechanically operated (see ). This method must be used as a referee method.
1.3 Although the mass and height of mass drop for each apparatus are the same, density achieved in compacted specimens with the same number of blows will be different. It is up to the owner or specifier to establish the specific required number of blows to be used for compaction of the specimen in relation to the field.
1.4 Units—The values stated in inch-pound units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
1.5 The text of this standard references notes and footnotes which provide explanatory material. These notes and footnotes (excluding those in tables and figures) shall not be considered as requirements of the standard.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D8 Terminology Relating to Materials for Roads and Pavements
D3666 Specification for Minimum Requirements for Agencies Testing and Inspecting Road and Paving Materials
D4402/D4402M Test Method for Viscosity Determination of Asphalt at Elevated Temperatures Using a Rotational Viscometer
D6927 Test Method for Marshall Stability and Flow of Asphalt Mixtures
E1 Specification for ASTM Liquid-in-Glass Thermometers
E11 Specification for Woven Wire Test Sieve Cloth and Test Sieves
E77 Test Method for Inspection and Verification of Thermometers
E2251 Specification for Liquid-in-Glass ASTM Thermometers with Low-Hazard Precision Liquids
ICS Number Code 93.080.20 (Road construction materials)
UNSPSC Code 30121500(Bituminous derivatives)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D6926-20, Standard Practice for Preparation of Asphalt Mixture Specimens Using Marshall Apparatus, ASTM International, West Conshohocken, PA, 2020, www.astm.orgBack to Top