| ||Format||Pages||Price|| |
|16||$58.00||  ADD TO CART|
|Hardcopy (shipping and handling)||16||$58.00||  ADD TO CART|
Significance and Use
5.1 This guide describes stabilization criteria for recording field measurements of temperature, DO, SC, and pH.
5.2 This guide describes the procedures used to calibrate and check meters to be used in the field to records these measurements and the procedures to be use in the field to obtain these data.
5.3 This guide describes quality assurance procedures to be followed when obtaining cross-sectional means of temperature, DO, SC, and pH of water flowing in open channels.
5.4 Field measurement must accurately represent the water flowing in the open channel being measured. Methods need to be used that will result in an accurate representation of the mean of the parameter of interest. Procedures must be used that will take into consideration the variation in the parameter across the sections and with depth.
5.5 Temperature and DO must be measured directly in the water in the open channel. SC and pH are often measured in situ, but also may be measured in a subsample of a composite sample collected using discharge-weighted methods.
1.1 This guide describes procedures to collect cross-sectional means of temperature, dissolved oxygen (DO), specific electrical conductance (SC), and pH of water flowing in open channels.
1.2 This guide provides guidelines for preparation and calibration of the equipment to collect cross-sectional means of temperature, DO, SC, and pH of water flowing in open channels.
1.3 This guide describes what equipment should be used to collect cross-sectional means of temperature, DO, SC, and pH of water flowing in open channels.
1.4 This guide covers the cross-sectional means of temperature, DO, SC, and pH of fresh water flowing in open channels.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D888 Test Methods for Dissolved Oxygen in Water
D1125 Test Methods for Electrical Conductivity and Resistivity of Water
D1129 Terminology Relating to Water
D1293 Test Methods for pH of Water
D4410 Terminology for Fluvial Sediment
D4411 Guide for Sampling Fluvial Sediment in Motion
D5464 Test Method for pH Measurement of Water of Low Conductivity
ICS Number Code 13.060.50 (Examination of water for chemical substances)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D6764-02(2019), Standard Guide for Collection of Water Temperature, Dissolved-Oxygen Concentrations, Specific Electrical Conductance, and pH Data from Open Channels, ASTM International, West Conshohocken, PA, 2019, www.astm.orgBack to Top