| ||Format||Pages||Price|| |
|5||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||5||$52.00||  ADD TO CART|
Significance and Use
5.1 The pour point of a petroleum product is an index of the lowest temperature of its utility for certain applications. Flow characteristics, like pour point, can be critical for the correct operation of lubricating systems, fuel systems, and pipeline operations.
5.2 Petroleum blending operations require precise measurement of the pour point.
5.3 Test results from this test method can be determined at either 1 °C or 3 °C intervals.
5.4 This test method yields a pour point in a format similar to Test Method /IP 15 when the 3 °C interval results are reported. However, when specification requires Test Method /IP 15, do not substitute this test method.
Note 2: Since some users may wish to report their results in a format similar to Test Method /IP 15 (in 3 °C intervals), the precision data were derived for the 3 °C intervals. For statements on bias relative to Test Method /IP 15, see .
5.5 This test method has better repeatability and reproducibility relative to Test Method /IP 15 as measured in the 1998 interlaboratory test program (see Section ).
1.1 This test method covers the determination of pour point of petroleum products by an automatic apparatus that applies a slightly positive air pressure onto the specimen surface while the specimen is being cooled.
1.2 This test method is designed to cover the range of temperatures from −57 °C to +51 °C; however, the range of temperatures included in the (1998) interlaboratory test program only covered the temperature range from −51 °C to −11 °C.
1.3 Test results from this test method can be determined at either 1 °C or 3 °C testing intervals.
1.4 This test method is not intended for use with crude oils.
Note 1: The applicability of this test method on residual fuel samples has not been verified. For further information on the applicability, refer to .
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D97 Test Method for Pour Point of Petroleum Products
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products
ICS Number Code 75.080 (Petroleum products in general)
UNSPSC Code 15101500(Petroleum and distillates)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D6749-02(2018), Standard Test Method for Pour Point of Petroleum Products (Automatic Air Pressure Method), ASTM International, West Conshohocken, PA, 2018, www.astm.orgBack to Top