| ||Format||Pages||Price|| |
|15||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||15||$52.00||  ADD TO CART|
|Standard + Redline PDF Bundle||30||$62.40||  ADD TO CART|
Significance and Use
5.1 The measurement of particulate matter and collected residue emission rates is an important test method widely used in the practice of air pollution control. Particulate matter measurements after control devices are necessary to determine total emission rates into the atmosphere.
5.1.1 These measurements, when approved by national, state, provincial, or other regional agencies, are often required for the purpose of determining compliance with regulations and statutes.
5.1.2 The measurements made before and after control devices are often necessary to demonstrate conformance with regulatory or contractual performance specifications.
5.2 The collected residue obtained with this test method is also important in characterizing stack emissions. However, the utility of these data is limited unless a chemical analysis of the collected residue is performed.
5.3 These measurements also can be used to calibrate continuous particulate emission monitoring systems by correlating the output of the monitoring instruments with the data obtained by using this test method.
1.1 This test method covers a method for the measurement of particulate matter (dust) concentration in emission gases in the concentrations below 20 mg/m3 standard conditions, with special emphasis around 5 mg/m3.
1.2 To meet the requirements of this test method, the particulate sample is weighed to a specified level of accuracy. At low dust concentrations, this is achieved by:
1.2.1 Precise and repeatable weighing procedures,
1.2.2 Using low tare weight weighing dishes,
1.2.3 Extending the sampling time at conventional sampling rates, or
1.2.4 Sampling at higher rates at conventional sampling times (high-volume sampling).
1.3 This test method differs from Test Method by requiring the mass measurement of filter blanks, specifying weighing procedures, and requiring monitoring of the flue gas flow variability over the testing period. It requires that the particulate matter collected on the sample filter have a mass at least five times a positive mass difference on the filter blank. High volume sampling techniques or an extension of the sampling time may be employed to satisfy this requirement. This test method has tightened requirements on sampling temperature fluctuations and isokinetic sampling deviation. This test method has eliminated the in-stack filtration technique.
1.4 This test method may be used for calibration of automated monitoring systems (AMS). If the emission gas contains unstable, reactive, or semi-volatile substances, the measurement will depend on the filtration temperature.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1193 Specification for Reagent Water
D1356 Terminology Relating to Sampling and Analysis of Atmospheres
D2986 Practice for Evaluation of Air Assay Media by the Monodisperse DOP (Dioctyl Phthalate) Smoke Test
D3154 Test Method for Average Velocity in a Duct (Pitot Tube Method)
D3631 Test Methods for Measuring Surface Atmospheric Pressure
D3670 Guide for Determination of Precision and Bias of Methods of Committee D22
D3685/D3685M Test Methods for Sampling and Determination of Particulate Matter in Stack Gases
D3796 Practice for Calibration of Type S Pitot Tubes
E1 Specification for ASTM Liquid-in-Glass Thermometers
E2251 Specification for Liquid-in-Glass ASTM Thermometers with Low-Hazard Precision Liquids
ISO StandardsISO5725 Precision of test methodsDetermination of repeatability and reproducibility by inter-laboratory tests ISO9096 Stationary source emissionsDetermination of concentration and mass flow rate of particulate material in gas-carrying ducts. Manual gravimetric method
ICS Number Code 13.040.99 (Other standards related to air quality)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D6331-16, Standard Test Method for Determination of Mass Concentration of Particulate Matter from Stationary Sources at Low Concentrations (Manual Gravimetric Method), ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top