| ||Format||Pages||Price|| |
|6||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$46.00||  ADD TO CART|
Significance and Use
5.1 The pour point of a petroleum product is an index of the lowest temperature of its utility for certain applications. Flow characteristics, such as pour point, can be critical for the correct operation of lubricating systems, fuel systems, and pipeline operations.
5.2 Petroleum blending operations require precise measurement of the pour point.
5.3 This test method can determine the temperature of the test specimen with a resolution of 0.1°C at which either crystals have formed or viscosity increases sufficiently to impede movement of the petroleum product.
5.4 This test method yields a pour point in a format similar to Test Method D97/IP15 when the 3°C interval results are reported.
5.5 This test method has better repeatability and comparable reproducibility relative to Test Method D97 as measured in the 1992 interlaboratory program. (See Section 13.)
1.1 This test method covers the determination of pour point of petroleum products by an automatic instrument that continuously rotates the test specimen against a suspended detection device during cooling of the test specimen.
1.2 This test method is designed to cover the range of temperatures from −57 to +51°C; however, the range of temperatures included in the 1992 interlaboratory program only covered the temperature range of − 39 to +6°C (see 13.4).
1.3 This test method determines the no-flow point of petroleum products by detection of the crystal structure or viscosity increase, or both, in the sample that is sufficient to impede flow of the specimen.
1.4 This test method is not intended for use with crude oils.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D97 Test Method for Pour Point of Petroleum Products
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products
Energy Institute StandardsIP 15 Test Method for Pour Point of Petroleum Products
ICS Number Code 75.080 (Petroleum products in general)
UNSPSC Code 15101500(Petroleum and distillates)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D5985-02(2014), Standard Test Method for Pour Point of Petroleum Products (Rotational Method), ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top