| ||Format||Pages||Price|| |
|3||$40.00||  ADD TO CART|
|Hardcopy (shipping and handling)||3||$40.00||  ADD TO CART|
Significance and Use
4.1 Due to the inherent danger of peroxides in butadiene, specification limits are usually set for their presence. This test method will provide values that can be used to determine the peroxide content of a sample of commercial butadiene.
4.2 Butadiene polyperoxide is a very dangerous product of the reaction between butadiene and oxygen that can occur. The peroxide has been reported to be the cause of some violent explosions in vessels that are used to store butadiene.
1.1 This test method covers the determination of peroxides in butadiene.
1.2 This test method covers the concentrations range of 1 to 10 ppm by mass (ppmw) as available oxygen.
1.3 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1265 Practice for Sampling Liquefied Petroleum (LP) Gases, Manual Method
D3700 Practice for Obtaining LPG Samples Using a Floating Piston Cylinder
ICS Number Code 71.080.10 (Aliphatic hydrocarbons)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D5799-95(2014), Standard Test Method for Determination of Peroxides in Butadiene, ASTM International, West Conshohocken, PA, 2014, www.astm.orgBack to Top