| ||Format||Pages||Price|| |
|10||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||10||$46.00||  ADD TO CART|
|Standard + Redline PDF Bundle||20||$55.20||  ADD TO CART|
Significance and Use
6.1 pH measurement of low conductivity water is frequently applied to power plant water and condensed steam samples for corrosion and scale prevention. It is sometimes used in pure water treatment systems between multiple pass membranes to optimize performance.
6.2 High purity water is highly unbuffered and small amounts of contamination can change the pH significantly. Specifically, high purity water rapidly absorbs CO2 gas from the atmosphere, which lowers the pH of the sample. The sample container and accompanying pH measurement technique minimize exposure of the high purity water sample to the atmosphere.
6.3 The high purity water sample may contain volatile trace components that will dissipate from the sample if exposed to the atmosphere. The sample container used in this test method will prevent these losses.
6.4 High purity water has a significant solution temperature coefficient. For greatest accuracy the sample to be measured should be close to the temperature of the sample stream and appropriate compensation should be applied.
6.5 When the preferred Test Method , which requires a real-time, flowing sample, cannot be utilized for practical reasons such as physical plant layout, unacceptable loss of water, location of on-line equipment sample points, or availability of dedicated test equipment, this method offers a viable alternative. The most significant difference between the two test methods is that Test Method obtains a real-time pH measurement from a flowing sample and this method obtains a time delayed pH measurement from a static grab sample.
6.6 pH measurements of low conductivity water are always subject to interferences ( ) and Test Method is more effective in eliminating these interferences especially with regard to contamination. This static grab sample method is more prone to contamination and temperature-induced errors because of the time lag between the sampling in the plant and sample pH reading which is taken in the laboratory.
1.1 This test method is applicable to determine the pH of water samples with a conductivity of 2 to 100 μS/cm over the pH range of 3 to 11 and is frequently used in power generation low conductivity samples. pH measurements of water of low conductivity are problematic. Specifically, this test method avoids contamination of the sample with atmospheric gases and prevents volatile components of the sample from escaping. This test method provides for pH electrodes and apparatus that address additional considerations discussed in . This test method also minimizes problems associated with the sample's pH temperature coefficient when the operator uses this test method to calibrate an on-line pH monitor or controller (see ).
1.2 This test method covers the measurement of pH in water of low conductivity with a lower limit of 2.0 μS/cm, utilizing a static grab-sample procedure where it is not practicable to take a real-time flowing sample.
Note 1: Test Method for on-line measurement is preferred over this method whenever possible. Test Method is not subject to the limited conductivity range, temperature interferences, potential KCl contamination, and time limitations found with this method.
1.3 For on-line measurements in water with conductivity of 100 μS/cm and higher, see Test Method .
1.4 For laboratory measurements in water with conductivity of 100 μS/cm and higher, see Test Method .
1.5 The values stated in SI units are to be regarded as standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D1293 Test Methods for pH of Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D3370 Practices for Sampling Water from Closed Conduits
D4453 Practice for Handling of High Purity Water Samples
D5128 Test Method for On-Line pH Measurement of Water of Low Conductivity
D6569 Test Method for On-Line Measurement of pH
ICS Number Code 13.060.50 (Examination of water for chemical substances)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D5464-16, Standard Test Method for pH Measurement of Water of Low Conductivity, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top