| ||Format||Pages||Price|| |
|7||$54.00||  ADD TO CART|
|Hardcopy (shipping and handling)||7||$54.00||  ADD TO CART|
Significance and Use
5.1 Fe-55 is formed in reactor coolant systems of nuclear reactors by activation of stable iron. The 55Fe is not completely removed by waste processing systems and some is released to the environment by means of normal waste liquid discharges. Power plants are required to monitor these discharges for 55Fe as well as other radionuclides.
5.2 This technique effectively removes other activation and fission products such as isotopes of iodine, zinc, manganese, cobalt, and cesium by the addition of hold-back carriers and an anion exchange technique. The fission products (zirconium-95 and niobium-95) are selectively eluted with hydrochloric-hydrofluoric acid washes. The iron is finally separated from Zn+2 by precipitation of FePO4 at a pH of 3.0.
1.1 This test method covers the determination of 55Fe in the presence of 59Fe by liquid scintillation counting. The a-priori minimum detectable concentration for this test method is 7.4 Bq/L.
1.2 This test method was developed principally for the quantitative determination of 55Fe. However, after proper calibration of the liquid scintillation counter with reference standards of each nuclide, 59Fe may also be quantified.
1.3 This test method was used successfully with Type III reagent water conforming to Specification . It is the responsibility of the user to ensure the validity of this test method for waters of untested matrices.
1.4 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For a specific hazard statement, see Section .
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1068 Test Methods for Iron in Water
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D3370 Practices for Sampling Water from Closed Conduits
D5847 Practice for Writing Quality Control Specifications for Standard Test Methods for Water Analysis
D7282 Practice for Set-up, Calibration, and Quality Control of Instruments Used for Radioactivity Measurements
ICS Number Code 13.060.50 (Examination of water for chemical substances); 13.060.60 (Examination of water for physical properties)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D4922-09(2016)e1, Standard Test Method for Determination of Radioactive Iron in Water, ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top