| ||Format||Pages||Price|| |
|6||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$46.00||  ADD TO CART|
Significance and Use
5.1 This test method is useful for analysis of many natural waters that contain thiocyanate from organic decomposition products and waste water discharges. Some industrial wastes, such as those from the metallurgical processing of gold ores, steel industry, petroleum refining, and coal gasification, may contain significant concentrations of thiocyanate. Thiocyanate per se is not recognized as a toxic chemical compound. However, when chlorinated, thiocyanate is converted to the highly toxic and volatile cyanogen chloride at high pH. Oxidation of thiocyanate may also release toxic hydrogen cyanide. The user of the method is advised to perform holding time studies in accordance with Practice D4841 whenever oxidants are present in the samples.
5.1.1 For information on the impact of cyanogens and cyanide compounds, see Appendix X1 of Test Method D2036.
1.1 This test method covers the quantitative colorimetric laboratory measurement of dissolved thiocyanate in water, waste water, and saline water in the range from 0.1 to 2.0 mg/L. For higher concentrations, use an aliquot from the diluted sample.
1.1.1 Validation—This test method was validated over the range of 0.07 to 1.42 mg/L. This test method was validated at nine laboratories at four levels. This test method may be valid for reporting results down to lower levels as validated in individual user laboratories.
1.1.2 Application—This test method has been validated in reagent water, Type II, in multiple laboratories and 7 natural waters, 1 laboratory effluent, 1 steel mill effluent, and 2 dechlorinated and treated sanitary effluents in single laboratories. It is the user's responsibility to assure the validity of the test method on any untested matrices.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. For specific hazard statements, see Section 9.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1129 Terminology Relating to Water
D1192 Guide for Equipment for Sampling Water and Steam in Closed Conduits
D1193 Specification for Reagent Water
D2036 Test Methods for Cyanides in Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D3370 Practices for Sampling Water from Closed Conduits
D3856 Guide for Management Systems in Laboratories Engaged in Analysis of Water
D4210 Practice for Intralaboratory Quality Control Procedures and a Discussion on Reporting Low-Level Data
D4841 Practice for Estimation of Holding Time for Water Samples Containing Organic and Inorganic Constituents
D5788 Guide for Spiking Organics into Aqueous Samples
D5789 Practice for Writing Quality Control Specifications for Standard Test Methods for Organic Constituents
D5847 Practice for Writing Quality Control Specifications for Standard Test Methods for Water Analysis
D7237 Test Method for Free Cyanide with Flow Injection Analysis (FIA) Utilizing Gas Diffusion Separation and Amperometric Detection
D7365 Practice for Sampling, Preservation and Mitigating Interferences in Water Samples for Analysis of Cyanide
E60 Practice for Analysis of Metals, Ores, and Related Materials by Spectrophotometry
E275 Practice for Describing and Measuring Performance of Ultraviolet and Visible Spectrophotometers
ICS Number Code 13.060.50 (Examination of water for chemical substances)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D4193-08(2013)e1, Standard Test Method for Thiocyanate in Water, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top