| ||Format||Pages||Price|| |
|6||$46.00||  ADD TO CART|
|Hardcopy (shipping and handling)||6||$46.00||  ADD TO CART|
Significance and Use
5.1 This test method was developed for measuring tritium in water to determine if the concentration exceeds the regulatory statutes of drinking water. This test method also is applicable for the determination of tritium concentration in water as required by technical specifications governing the operations of nuclear power facilities. With suitable counting technique, sample size, and counting time a detection limit of less than 37 Bq/L (1000 pCi/L) is attainable by liquid scintillation.
1.3 The tritium concentrations, which can be measured by this test method utilizing currently available liquid scintillation instruments, range from less than 0.037 Bq/mL (1 pCi/mL) to 555 Bq/mL (15 000 pCi/mL) for a 10-mL sample aliquot. Higher tritium concentrations can be measured by diluting or using smaller sample aliquots, or both.
1.4 The maximum contaminant level for tritium in drinking water as given by the United States Environmental Protection Agency (U.S. EPA) National Interim Primary Drinking Water Regulations (NIPDWR) is 0.740 Bq/mL (20 pCi/mL). The NIPDWR lists a required detection limit for tritium in drinking water of 0.037 Bq/mL (1 pCi/mL), meaning that drinking water supplies, where required, should be monitored for tritium at a sensitivity of 0.037 Bq/mL (1 pCi/mL). In Appendix X1, Eq X1.3 is given for determining the necessary counting time to meet the required sensitivity for drinking water monitoring.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D1129 Terminology Relating to Water
D1193 Specification for Reagent Water
D2777 Practice for Determination of Precision and Bias of Applicable Test Methods of Committee D19 on Water
D3370 Practices for Sampling Water from Closed Conduits
D3648 Practices for the Measurement of Radioactivity
ICS Number Code 13.060.50 (Examination of water for chemical substances)
UNSPSC Code 50202301(Water)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D4107-08(2013), Standard Test Method for Tritium in Drinking Water, ASTM International, West Conshohocken, PA, 2013, www.astm.orgBack to Top