| ||Format||Pages||Price|| |
|13||$60.00||  ADD TO CART|
|Hardcopy (shipping and handling)||13||$60.00||  ADD TO CART|
Significance and Use
4.1 The water and sediment content of crude oil is significant because it can cause corrosion of equipment and problems in processing. A determination of water and sediment content is required to measure accurately net volumes of actual oil in sales, taxation, exchanges, and custody transfers. It is not anticipated that this test method, which is written with a dedicated laboratory facility in mind, is likely to be used in field test rooms or sample rooms due to safety concerns for proper ventilation and handling.
4.2 This test method may not be suitable for crude oils that contain alcohols that are soluble in water. In cases where the impact on the results may be significant, the user is advised to consider using another test method, such as Test Method (API MPMS Chapter 10.9).
1.1 This test method describes the laboratory determination of water and sediment in crude oils by means of the centrifuge procedure. This centrifuge method for determining water and sediment in crude oils is not entirely satisfactory. The amount of water detected is almost always lower than the actual water content. When a highly accurate value is required, the revised procedures for water by distillation, Test Method (API MPMS Chapter 10.2) ( ), and sediment by extraction, Test Method (API MPMS Chapter 10.1), shall be used.
Note 1: Test Method MPMS Chapter 10.2) has been determined to be the preferred and most accurate method for the determination of water. (API
1.2 The values stated in SI units are to be regarded as standard.
1.2.1 Exception—The values given in parentheses are for information only.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Specific warning statements appear in , , and .
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
D95 Test Method for Water in Petroleum Products and Bituminous Materials by Distillation (API MPMS Chapter 10.5)
D473 Test Method for Sediment in Crude Oils and Fuel Oils by the Extraction Method (API MPMS Chapter 10.1)
D665 Test Method for Rust-Preventing Characteristics of Inhibited Mineral Oil in the Presence of Water
D4006 Test Method for Water in Crude Oil by Distillation (API MPMS Chapter 10.2)
D4057 Practice for Manual Sampling of Petroleum and Petroleum Products (API MPMS Chapter 8.1)
D4177 Practice for Automatic Sampling of Petroleum and Petroleum Products (API MPMS Chapter 8.2)
D4928 Test Method for Water in Crude Oils by Coulometric Karl Fischer Titration (API MPMS Chapter 10.9)
D5854 Practice for Mixing and Handling of Liquid Samples of Petroleum and Petroleum Products (API MPMS Chapter 8.3)
E969 Specification for Glass Volumetric (Transfer) Pipets
ICS Number Code 75.040 (Crude petroleum)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM D4007-11(2016)e1, Standard Test Method for Water and Sediment in Crude Oil by the Centrifuge Method (Laboratory Procedure), ASTM International, West Conshohocken, PA, 2016, www.astm.orgBack to Top