| ||Format||Pages||Price|| |
|4||$40.00||  ADD TO CART|
|Hardcopy (shipping and handling)||4||$40.00||  ADD TO CART|
|Standard + Redline PDF Bundle||8||$48.00||  ADD TO CART|
Significance and Use
5.1 Uranium hexafluoride is a basic material used to prepare nuclear reactor fuel. To be suitable for this purpose, the material shall meet the criteria for isotopic composition. This test method is designed to determine whether the material meets the requirements described in Specifications and .
5.2 Fitness for Purpose of Safeguards and Nuclear Safety Applications—Methods intended for use in safeguards and nuclear safety applications shall meet the requirements specified by Guide for use in such applications.
5.2.1 The material (uranium hexafluoride) to which this test method applies is subject to the nuclear safeguards regulations governing its possession and use. The analytical procedure in this test method has been designated as technically acceptable for generating safeguards accountability data.
5.2.2 When used in conjunction with the appropriate certified reference materials (CRMs), this procedure can demonstrate traceability to the national measurement base. However, adherence to this procedure does not automatically guarantee regulatory acceptance of the regulatory safeguards measurements. It remains the sole responsibility of the user of this test method to ensure that its application to safeguards has the approval of the proper regulatory authorities.
1.1 This is a quantitative test method applicable to determining the mass percent of uranium isotopes in uranium hexafluoride (UF6) samples with 235U concentrations between 0.1 and 5.0 mass %.
1.2 This test method may be applicable for the entire range of 235U concentrations for which adequate standards are available.
1.3 This test method is for analysis by a gas magnetic sector mass spectrometer with a single collector using interpolation to determine the isotopic concentration of an unknown sample between two characterized UF6 standards.
1.4 This test method is to replace the existing test method currently published in Test Methods and is used in the nuclear fuel cycle for UF6 isotopic analyses.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
C761 Test Methods for Chemical, Mass Spectrometric, Spectrochemical, Nuclear, and Radiochemical Analysis of Uranium Hexafluoride
C787 Specification for Uranium Hexafluoride for Enrichment
C996 Specification for Uranium Hexafluoride Enriched to Less Than 5 %235U
Other DocumentUSEC-651 Uranium Hexafluoride: A Manual of Good Handling Practices, Latest Revisions
ICS Number Code 27.120.30 (Fissile materials and nuclear fuel technology)
UNSPSC Code 30111500(Concrete and mortars)
UNSPSC Code 15131500(Nuclear fuel)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1742-17, Standard Test Method for Isotopic Analysis of Uranium Hexafluoride by Double Standard Single-Collector Gas Mass Spectrometer Method, ASTM International, West Conshohocken, PA, 2017, www.astm.orgBack to Top