| ||Format||Pages||Price|| |
|7||$52.00||  ADD TO CART|
|Hardcopy (shipping and handling)||7||$52.00||  ADD TO CART|
Significance and Use
5.1 The following methods assist in demonstrating regulatory compliance in such areas as safeguards (Special Nuclear Material), inventory control, criticality control, decontamination and decommissioning, waste disposal, holdup and shipping.
5.2 This guide can apply to the assay of radionuclides in containers, whose gamma-ray absorption properties can be measured or estimated, for which representative certified standards are not available. It can be applied to in situ measurements, measurement stations, or to laboratory measurements.
5.3 Some of the modeling techniques described in the guide are suitable for the measurement of fall-out or natural radioactivity homogenously distributed in soil.
5.4 Source-based efficiency calibrations for laboratory geometries may suffer from inaccuracies due to gamma rays being detected in true coincidence. Modeling can be an advantage since it is unaffected by true coincidence summing effects.
1.1 This guide addresses the use of models with passive gamma-ray measurement systems. Mathematical models based on physical principles can be used to assist in calibration of gamma-ray measurement systems and in analysis of measurement data. Some nondestructive assay (NDA) measurement programs involve the assay of a wide variety of item geometries and matrix combinations for which the development of physical standards are not practical. In these situations, modeling may provide a cost-effective means of meeting user’s data quality objectives.
1.2 A scientific knowledge of radiation sources and detectors, calibration procedures, geometry and error analysis is needed for users of this standard. This guide assumes that the user has, at a minimum, a basic understanding of these principles and good NDA practices (see Guide ), as defined for an NDA professional in Guide . The user of this standard must have at least a basic understanding of the software used for modeling. Instructions or further training on the use of such software is beyond the scope of this standard.
1.3 The focus of this guide is the use of response models for high-purity germanium (HPGe) detector systems for the passive gamma-ray assay of items. Many of the models described in this guide may also be applied to the use of detectors with different resolutions, such as sodium iodide or lanthanum halide. In such cases, an NDA professional should determine the applicability of sections of this guide to the specific application.
1.4 Techniques discussed in this guide are applicable to modeling a variety of radioactive material including contaminated fields, walls, containers and process equipment.
1.5 This guide does not purport to discuss modeling for “infinite plane” in situ measurements. This discussion is best covered in ANSI N42.28.
1.6 This guide does not purport to address the physical concerns of how to make or set up equipment for in situ measurements but only how to select the model for which the in situ measurement data is analyzed.
1.7 The values stated in either SI units or inch-pound units are to be regarded separately as standard. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Combining values from the two systems may result in non-conformance with the standard.
1.8 The values stated in inch-pound units are to be regarded as standard. The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
1.9 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
2. Referenced Documents (purchase separately) The documents listed below are referenced within the subject standard but are not provided as part of the standard.
Other StandardANSI N42.28 Performance Standard for the Calibration of Germanium Detectors for In Situ Gamma-Ray Measurements
C1490 Guide for the Selection, Training and Qualification of Nondestructive Assay (NDA) Personnel
C1673 Terminology of C26.10 Nondestructive Assay Methods
ICS Number Code 17.240 (Radiation measurements)
UNSPSC Code 41111929(Radiation detectors)
|Link to Active (This link will always route to the current Active version of the standard.)|
ASTM C1726 / C1726M-10(2018), Standard Guide for Use of Modeling for Passive Gamma Measurements, ASTM International, West Conshohocken, PA, 2018, www.astm.orgBack to Top